Measurement SystemNextEngine proprietary MultiStripe Laser Triangulation (MLT) technology. Patents Pending.
SourceTwin arrays of four, Class 1M, 10 mW solid state lasers with custom optics 650 nm λ.
SensorTwin 5.0 Megapixel CMOS image sensors.
Photo SurfaceOptically synchronous 7-color surface capture for precision-locked geomtry correlation.
Photo LightingBuilt-in spatially diverse whitelight illuminators with tri-phosphor, wide color gamut.
AutoDrive™High precision rotary servo positioner, auto-controlled by scanner. 20 lb max.
PartGripper™Universal part holder to adjust height, angle, and orientation of capture. 10 lb capacity.
Object SizeNo preset limit. Objects larger than field can be assembled with supplied software.
Field Size5.1" x 3.8" (Macro) and 13.5" x 10.1" (Wide). ProScan Extended Mode 22.5” x 16.75”.
Capture DensityOn target surface capture up to 268K points/in²(Marco), 29K points/in² (Wide).
Texture Density500 DPI on target surfaces is 500 DPI in Macro Mode and 200 DPI in Wide Mode.
Dimensional Accuracy±0.005" in Macro Mode and ±0.015" in Wide Mode.
Acquisition Speed50,000 processed points/sec throughput. Typically 2 minutes per scan of each facet.
Typical DatasetsTypical small models are a quarter million points, after oversampling and optimization
EnvironmentalDesktop use under ordinary office lighting. No darkroom or special background required.