|Publication number||US6533751 B2|
|Application number||US 09/758,331|
|Publication date||18 Mar 2003|
|Filing date||9 Jan 2001|
|Priority date||9 Jan 2001|
|Also published as||CA2423604A1, DE60133410D1, DE60133410T2, EP1349486A1, EP1349486A4, EP1349486B1, US20020091372, WO2002065897A1|
|Publication number||09758331, 758331, US 6533751 B2, US 6533751B2, US-B2-6533751, US6533751 B2, US6533751B2|
|Inventors||Andrew Cragg, James H. Kim|
|Original Assignee||Andrew Cragg, James H. Kim|
|Export Citation||BiBTeX, EndNote, RefMan|
|Patent Citations (11), Referenced by (35), Classifications (21), Legal Events (4)|
|External Links: USPTO, USPTO Assignment, Espacenet|
This invention relates intravascular catheters, and more particularly to micro catheters and guidewires that access tortuous regions of the vasculature.
There is a demand for increasingly smaller diameter catheters to enable access to tortuous regions of the vascular system such as regions of the neurovasculature.
Tortuous regions of the vasculature are defined as regions having vessels that branch off from more proximal vessels at angles of greater than 90 degrees. Portions of the vessels have lumen diameters of 3 mm or less. Micro catheters are defined as those catheters capable of navigating through these tortuous regions. There are limitations to the functionality of existing micro catheters.
One limitation relates to pushability of micro catheters. Typically micro catheters are inserted into the vasculature with a guiding catheter. As micro catheters evolve into smaller sizes, it is found that a micro catheter can kink or buckle when tracking via the guiding catheter. Kinking and buckling of a micro catheter are not desirable qualities.
Many micro catheters have a braided catheter body to reinforce the catheter body, optimizing catheter pushability and thereby inhibiting buckling. Current braided catheter technology has proved useful with the larger micro catheters; e.g. micro catheters having a 0.032″ distal shaft diameter and larger.
Braided catheters, particularly for micro catheters having distal shaft diameters of less than 0.032″, are expensive to manufacture and may be cost prohibitive to use regularly. Braided catheters may not bend well enough for use in the most distal and tortuous regions of the vasculature. What is desired is a micro catheter that has a high degree of axial compressive strength (pushability) and the capability to bend through tortuous regions of the vasculature. What is also desired is a micro catheter that resists kinking and buckling.
A micro catheter and guidewire system for use in tortuous regions of a vasculature includes a catheter body having a distal end, a guidewire lumen, a proximal region with an outer diameter, and a distal region with an outer diameter.
The ratio of the proximal region outer diameter to the distal region outer diameter is at least 1.625:1 to strengthen the proximal region, thereby improving the overall integrity of the catheter body. Increased torque capability, improved pushability and increased control of the catheter body result from forming the catheter body according to this ratio. Strengthening the proximal region further enables the catheter body to resist kinking and buckling during use.
The guidewire lumen defines a stopper and the guidewire has a depth stop for engaging the stopper of the guidewire lumen. The stopper in the guidewire lumen and depth stop on the guidewire enable the guidewire to selectively and compositely cooperate with the catheter to improve catheter pushability when the depth stop and the stopper engage.
The depth stop and stopper cooperate to prevent over extension of the guidewire when the guidewire and catheter body simultaneously push through tortuous regions of the vasculature.
The guidewire lumen defines a pathway between the guidewire and the guidewire lumen to facilitate infusion of fluids including contrast agents, blood thinners, nutrients, and medicine through the distal end of the catheter body. This is important because a separate infusion lumen, which consumes space, is not necessarily required. The depth stop and stopper function as a valve to prevent the infusion of fluid via the pathway.
The present invention is described by way of example in the following figures where like parts have like reference numerals and wherein:
FIG. 1 shows a catheter and guidewire in accordance with the present invention.
FIG. 2a shows the catheter of FIG. 1 inserted into a patient.
FIG. 2b shows an expanded view of a portion of FIG. 2a
FIG. 3 shows an embodiment of a guidewire in accordance with the present invention.
FIG. 4 shows the distal end of a catheter body in accordance with the present invention.
FIG. 5 shows the guidewire of FIG. 3 within the catheter body of FIG. 4.
FIG. 6 shows an embodiment of a guidewire in accordance with the present invention.
FIG. 1 shows an intravascular catheter, generally designated with the reference numeral 10. The catheter 10 includes a guidewire 12 extending through the catheter 10. The catheter 10 includes a hollow catheter body 14 having a proximal region 16, a first intermediate region 18, a second intermediate region 20, and a distal region 22.
The distal region 22 includes marker bands 24 to facilitate identification of the distal region 22 when the catheter inserts into the vasculature of a patient. The catheter body has a distal end 26 defined on the distal region 24. The guidewire 12 extends from the distal end 26 when the guidewire 12 inserts through the catheter 10.
The catheter 10 can be designed to have any of a number of therapeutic or diagnostic functions. Preferably, the catheter body 14 includes an infusion lumen with ports 28 that facilitate direct delivery of fluids to the blood stream of a patient. Such fluids include nutrients, blood thinners, medicine, contrast agent, or other fluid useful in diagnosis and treatment of the patient.
The catheter body 14 is tubular, having a circular cross section. Each region 16, 18, 20, and 22 forms a discrete segment, having an outer diameter d1, d2, d3, and d4, respectively. The proximal region is bolstered in size and integrity with respect to the distal region. The ratio of the proximal region outer diameter d1 to the distal region outer diameter d4 is at least 1.625:1.
Bolstering the proximal region 16 improves not only the local strength of the proximal region 16, but importantly, improves the overall strength of the catheter body 14. Improving the overall strength of the catheter body 14 by bolstering the proximal region 16 results in measurable improvements in catheter 10 pushability, backup ability, torque transfer capability and control. Bolstering the proximal region 16 makes the distal region 22 resist kinking or buckling while accessing tortuous regions of the vasculature.
It can be appreciated that while bolstering the proximal region 16 is accomplished, according to the present invention, by increasing the proximal region diameter d1, there are other ways of bolstering the proximal region 16. Such ways include providing a proximal region of a relatively stronger material (as compared with the material of the distal region), or otherwise reinforcing the proximal region 16. Increasing the proximal region diameter d1 is preferred to these other techniques because it is simpler to accomplish, and it works.
An optimal catheter design, according to the present invention, has the ratio of the proximal region 16 outer diameter d1 to the distal region 22 outer diameter d4 of at least 1.625:1. This geometry proves useful alone, or in combination with using materials of varying strengths and flexibility for each of the regions 16, 18, 20 and 22. Braided reinforcement is used to further strengthen selected segments of the catheter body 14, as necessary.
Preferably, the catheter body 14 is constructed of thermoplastic materials where the proximal region 16 has the most rigid composition, and the distal region 22 has the softest composition. The composition of the intermediate regions 18 and 20 results in decreasing catheter body 14 stiffness between the proximal region and the distal region.
It can be appreciated that while two intermediate regions 18 and 20 are shown it is possible, and perhaps desirable, to have more, or less, intermediate regions. Further, while the intermediate regions 18 and 20 form discrete segments, the catheter body 14 can also taper from the proximal region 16 to the distal region 22 without the need for discrete segments. Design considerations such as catheter length, and particular application can dictate the appropriate number of intermediate regions, geometry, flexibility and composition of such regions.
A prototype of the invention tested to have regions of relative stiffness as follows: the proximal region 16 has a stiffness of 0.004-0.0020 in/lb at 30 degree deflection at span of 0.05″ with a 0.025 lb weight; the first intermediate region 18 has a stiffness of 0.0020-0.0080 in/lb at 30 degree deflection at span of 0.105″ with a 0.025 lb weight deflection; the second intermediate region 20 has a stiffness of 0.0020-0.0040 in/lb at 30 degree deflection at span of 0.05″ with a 0.080 lb weight; and the distal region 22 has a stiffness of 0.0030-0.0070 in/lb at 30 degree deflection at span of 0.05″ with a 0.080 lb weight. It can be appreciated that while these figures are reflective of stiffness values for a particular micro catheter 10 in accordance with the present invention (i.e. a micro catheter 10 having a distal region outer diameter of 0.032″ or less) these stiffness figures for micro catheter designs of very small sizes may be softer than these exemplarily stiffness ranges.
While the stiffness of each region of the catheter body 14 can be measured under lab conditions, there are other ways of representing the relative stiffness of the various regions of the catheter body 14, such as in terms of material composition and geometry.
One preferable geometry of the catheter body 14 has a usable length of between 120-180 cm. The proximal region 16 has a 3.0F-5.0F outer diameter d1, and an 80-130 cm length. More preferably, the proximal region 16 has a 4.0F-5.0F (0.052″-0.065″) outer diameter d1. The first intermediate region 18 has a 2.5F-5F outer diameter d2 and a 2-10 cm length. The second intermediate 20 region has a 2.0F-4.5F outer diameter d3 and a 5-20 cm length.
The distal region 22 has a 1.2F-2.5F outer diameter d4 and a 5-20 cm length. It can be appreciated that the distal end 26 of the catheter body 14 can be contoured, stepped or otherwise deviate from the average dimension d4. Accordingly, the d4 value generally referred to herein is the nominal dimension of the whole 5-20 cm length of the distal region 22 without regard to contours, steps or other deviations that may be present on the distal region 22, particularly near the distal end 26.
The distal region outer diameter d4 is a maximum of 2.5F (0.032″) to enable use of the catheter body in tortuous regions of the vasculature and the proximal region has a minimum outer diameter d1, of 4.0F (0.052″) to optimize catheter body pushability.
According to one aspect of the invention, the catheter body 14 includes an infusion lumen and infusion ports 28. Preferably, the infusion ports 28 are located in a transition region defined between the second intermediate region 20 and the distal region 22. While FIG. 1 shows infusion ports 28 at a single location, it can be appreciated that any transition region between discrete catheter segments may include infusion ports 28 in accordance with the present invention.
FIG. 2a and FIG. 2b show the micro catheter 10 inserted into the vasculature 36 of a patient 38. A method of using the catheter 10 in tortuous regions of the vasculature 36 includes providing the catheter body 14, disposing a guidewire 12 into the catheter body 14 and extending the guidewire 12 beyond the distal end 26 of the catheter 10. The method includes the steps of simultaneously advancing the guidewire 12 and catheter body 14 into the vasculature 36 of the patient 38. The method step of limiting the extension of the guidewire 12 with a depth stop prevents over extension of the guidewire 12.
Preventing over extension of the guidewire 12 prevents the guidewire 12 from unnecessarily disrupting the compliant tissues of the vasculature 36. Limiting the extension of the guidewire 12 enables the guidewire to help “push” the catheter body 14 through the vasculature 36.
The method includes step of infusing fluid through the catheter body 14, via the infusion port 28, into the vasculature. The step of advancing the catheter 10 includes advancing the catheter 10 into tortuous regions of the neurovascualture.
The method further includes the step of detecting the marker bands 24 (FIG. 1) to determine where the distal end 26 is located within the vasculature 35.
The catheter body 14 has a guidewire lumen (supra) defining a pathway that functions as an infusion lumen. The method further comprises the step of delivering contrast media, medicine, nutrition, or a blood thinning agent through the distal end 26 of the catheter body 14 via the pathway.
FIG. 3 shows a distal end 40 of the guidewire 12. The distal end 40 has a depth stop 42 for preventing the guidewire from over-extending from the catheter. According to one aspect of the invention, the depth stop 42 defines an annular flange 44 that aligns coaxially with the distal end 40. It can be appreciated that though an annular flange 44 is shown for preventing over-extension of the guidewire, the depth stop 42 can take other shapes. For example, the depth stop 42 can take the shape of a generally spherical bulb, a series of ridges, a frustum aligned coaxially on the distal end, or any other shape that would prevent over-extension.
FIG. 4 shows the distal region 22 of the catheter body 14. The distal region 22 includes a stopper 50 that selectively engages the depth stop 42 of the guidewire 12. The stopper 50 defines annular flange 52 that meets the guidewire depth stop 42 annular flange 44 to prevent over extension of the guidewire 12 through the distal end 26 of the catheter body 14 (FIG. 1).
The catheter body 14 defines a guidewire lumen 54 within the catheter body 14. The stopper 50 is fixed within the guidewire lumen 54 and has an opening to permit a guidewire to pass through the stopper 50.
With reference to FIG. 1, FIG. 2a, FIG. 2b, FIG. 3 and FIG. 4, it can be appreciated that pressing the guidewire depth stop 42 against the stopper 50 of the catheter body 14 improves the catheter body 14 pushability and inhibits buckling and kinking of the distal region 22 of the catheter body 14. This is possible because when the guidewire 12 (FIG. 1) fully extends from the catheter body 14, and the depth stop 42 presses against the stopper 50, the guidewire 12 and the catheter body 14 act in composite cooperation. Simultaneous insertion of the guidewire 12 and the catheter body 14 into the tortuous regions of the neuro-vasculature is enabled because the guidewire 12 reinforces the catheter body 14, particularly the distal region 22 of the catheter body 14.
Pressing an annular depth stop 42 against and an annular stopper 50 inhibits deformation, (e.g. expansion) of the outer diameter d4 of the distal region 22. However, it can be appreciated that various other depth stop 42 and stopper 50 designs would inhibit such deformation. It should also be noted that a limited degree of deformation is not always objectionable.
FIG. 5 shows the distal region 22 of the catheter body 14. The guidewire 12 extends through the guidewire lumen 54. The catheter body 14 forms a pathway 56 within the guidewire lumen 54, between the guidewire 12 and the guidewire lumen 54. The pathway 56 facilitates infusion of fluid through the distal end 26, which functions as an infusion port.
The guidewire lumen 54 has a stepped interior 58. The guidewire 12 has a stepped exterior 60 that corresponds with the stepped interior 58 of the guidewire lumen 54. The stepped exterior 60 is offset from the stepped interior 58 when the depth stop 42 of the guidewire 12 meets the stopper 50 of the catheter body 14. Offsetting the stepped exterior of the guidewire with the stepped interior of the depth stop 42 prevents the guidewire 12 from radially deforming the distal region 22 and the distal end 56 of the catheter body 14.
FIG. 6 shows an embodiment of the guidewire 12. The depth stop 42 includes a bulb 42 formed on the guidewire 12.
The present invention is described in terms of a preferred embodiment, however, it can be appreciated that the present invention can be modified to achieve various goals. For example, the ability of the guidewire to act as a valve can enable selective pressurization of the catheter body by infusion fluids to facilitate selective changes in catheter flexibility. These pressure changes in conjunction with selective reinforcement of the catheter body by the guidewire, and bolstering of the proximal region by the geometry and ratio described herein can result in smaller micro catheters with improved pushability. Further modifications to the configuration and ratio between the proximal region and distal region can also improve pushability in small micro catheters. The configuration of the stopper and the depth stop can be modified, and improved so that the guidewire can add more to the pushability of the catheter. Accordingly, the present invention is to be limited only by the following claims:
|Cited Patent||Filing date||Publication date||Applicant||Title|
|US4739768||2 Jun 1986||26 Apr 1988||Target Therapeutics||Catheter for guide-wire tracking|
|US5304198||13 Nov 1992||19 Apr 1994||Target Therapeutics||Single-lumen balloon catheter having a directional valve|
|US5318529||9 Aug 1991||7 Jun 1994||Boston Scientific Corporation||Angioplasty balloon catheter and adaptor|
|US5348537||15 Jul 1992||20 Sep 1994||Advanced Cardiovascular Systems, Inc.||Catheter with intraluminal sealing element|
|US5425723 *||30 Dec 1993||20 Jun 1995||Boston Scientific Corporation||Infusion catheter with uniform distribution of fluids|
|US5823198 *||31 Jul 1996||20 Oct 1998||Micro Therapeutics, Inc.||Method and apparatus for intravasculer embolization|
|US5873865 *||7 Feb 1997||23 Feb 1999||Eclipse Surgical Technologies, Inc.||Spiral catheter with multiple guide holes|
|US5888436 *||1 Jul 1997||30 Mar 1999||Hv Technologies, Inc.||Manufacture of variable stiffness microtubing|
|US5951494 *||16 Jan 1997||14 Sep 1999||Boston Scientific Corporation||Polymeric implements for torque transmission|
|US6179809 *||18 Sep 1998||30 Jan 2001||Eclipse Surgical Technologies, Inc.||Drug delivery catheter with tip alignment|
|US6312374 *||29 Sep 1999||6 Nov 2001||Progenix, Llc||Radioactive wire placement catheter|
|Citing Patent||Filing date||Publication date||Applicant||Title|
|US7022124 *||11 Apr 2002||4 Apr 2006||Pentax Corporation||Surgical instrument|
|US7601138 *||18 Jun 2002||13 Oct 2009||B. Braun Melsungen Ag||Catheter|
|US7704245||13 Apr 2004||27 Apr 2010||Cook Incorporated||Large diameter delivery catheter/sheath|
|US7763196||19 Oct 2007||27 Jul 2010||B. Braun Melsungen Ag||Catheter|
|US7780611||1 May 2003||24 Aug 2010||Boston Scientific Scimed, Inc.||Medical instrument with controlled torque transmission|
|US7815623||5 Oct 2005||19 Oct 2010||Genzyme Corporation||Stepped cannula|
|US7909873||14 Dec 2007||22 Mar 2011||Soteira, Inc.||Delivery apparatus and methods for vertebrostenting|
|US7959584 *||29 May 2002||14 Jun 2011||Boston Scientific Scimed, Inc.||Dedicated distal protection guidewires|
|US7959634||28 Mar 2005||14 Jun 2011||Soteira Inc.||Orthopedic surgery access devices|
|US7968038||8 Mar 2010||28 Jun 2011||Cook Medical Technologies Llc||Large diameter delivery catheter/sheath|
|US8292829||13 Jul 2010||23 Oct 2012||Boston Scientific Scimed, Inc.||Medical instrument with controlled torque transmission|
|US8337458||25 Dec 2012||Genzyme Corporation, A Sanofi Company||Stepped cannula|
|US8467854 *||20 Apr 2005||18 Jun 2013||Scimed Life Systems, Inc.||Neurovascular intervention device|
|US8608703||30 Oct 2007||17 Dec 2013||Medrad, Inc.||Infusion flow guidewire system|
|US8623025||15 Jan 2010||7 Jan 2014||Gmedelaware 2 Llc||Delivery apparatus and methods for vertebrostenting|
|US8821478||2 Mar 2012||2 Sep 2014||Boston Scientific Scimed, Inc.||Catheter with variable stiffness|
|US8845552||14 Sep 2012||30 Sep 2014||Boston Scientific Scimed, Inc.||Medical instrument with controlled torque transmission|
|US9089667||21 Aug 2006||28 Jul 2015||The Regents Of The University Of California||Reflux resistant cannula and system for chronic delivery of therapeutic agents using convection-enhanced delivery|
|US9192397||17 Jun 2009||24 Nov 2015||Gmedelaware 2 Llc||Devices and methods for fracture reduction|
|US9237916||14 Dec 2007||19 Jan 2016||Gmedeleware 2 Llc||Devices and methods for vertebrostenting|
|US9238119||11 Aug 2011||19 Jan 2016||Boston Scientific Limited||Infusion flow system and fluid coupling|
|US20020151903 *||11 Apr 2002||17 Oct 2002||Asahi Kogaku Kogyo Kabushiki Kaisha||Surgical instrument|
|US20030018317 *||18 Jun 2002||23 Jan 2003||Udo Goebel||Catheter|
|US20030225418 *||29 May 2002||4 Dec 2003||Scimed Life Systems, Inc.||Dedicated distal protection guidewires|
|US20040220499 *||1 May 2003||4 Nov 2004||Scimed Life Systems, Inc.||Medical instrument with controlled torque transmission|
|US20040220549 *||13 Apr 2004||4 Nov 2004||Dittman Jay A.||Large diameter delivery catheter/sheath|
|US20050090802 *||26 Apr 2004||28 Apr 2005||Connors John J.Iii||Flexible sheath with varying durometer|
|US20050209559 *||18 Mar 2005||22 Sep 2005||Medtronic Vascular, Inc. A Delaware Corporation||Apparatus and methods for the treatment of chronic total occlusions|
|US20050216018 *||28 Mar 2005||29 Sep 2005||Sennett Andrew R||Orthopedic surgery access devices|
|US20060135945 *||5 Oct 2005||22 Jun 2006||Avigen, Inc., The Regents Of The University Of California||Stepped cannula|
|US20060253023 *||20 Apr 2005||9 Nov 2006||Scimed Life Systems, Inc.||Neurovascular intervention device|
|US20080045927 *||19 Oct 2007||21 Feb 2008||B. Braun Melsungen Ag||Catheter|
|US20080058839 *||6 Feb 2007||6 Mar 2008||Nobles Anthony A||Reverse tapered guidewire and method of use|
|US20080312671 *||30 Oct 2007||18 Dec 2008||Possis Medical, Inc.||Infusion flow guidewire system|
|US20100163159 *||8 Mar 2010||1 Jul 2010||Cook Incorporated||Large diameter delivery catheter/sheath|
|U.S. Classification||604/93.01, 604/533, 604/96.01, 604/525, 604/264, 604/508, 604/524, 604/537, 604/506, 604/528, 604/507|
|International Classification||A61M25/18, A61M25/00, A61M25/16, A61M25/01|
|Cooperative Classification||A61M2025/0042, A61M2025/09083, A61M25/00, A61M2025/0063, A61M2025/09175|
|28 Feb 2003||AS||Assignment|
|13 Sep 2006||FPAY||Fee payment|
Year of fee payment: 4
|10 Sep 2010||FPAY||Fee payment|
Year of fee payment: 8
|18 Sep 2014||FPAY||Fee payment|
Year of fee payment: 12