Lenord+Bauer Rail Speed Sensor & Rotary Pulse Encoder Accessories
Mounting brackets, Installation, Condensed-water outlet
1) for all encoders GEL ... (excluding GEL 260 EEx)
Clamps, Flange, Measuring Wheel, Collet chuck
Measuring pedestal with slip free belt drive for incremental or absolute measurement of large distance for example on cranes.
Dimension: width 10 mm, tooth pitch 5 mm
Construction: 10 Steel-Cord strands (0.32 mm diameter bedded into Polyurethane-Elastomer)
Techn. Data: operating temperature - 10° C ... +80° C
short term 120° C
breaking load 1200 N
max. tension 300 N at 4 %o elasticity
Weight: 0.022 kg
Accuracy*: typ. ± 0.3 mm/m at 60 N toothed belt tension
In a mounted condition the accuracy can be improved by altering the tension. The stretching characteristic is linear. The coefficient is 0.01 mm / (1 m · 1 N).
Advantages: Unaffected by damp, petrol, oil, fats, hydrosol, uv-rays, ozone etc. Steel-Cord as tensioning.
strands: constant lengths at specified load, no additional stretching. (toothed belt for -30° C on request)
Protective casing SG 110/111
This casing protects rotary pulse encoders GEL 207/260 or absolute encoders GEL 152/153 against high mechanical stress and environmental conditions.Attention: The shaft of rotary pulse encoder GEL 207 must have a diameter of 12 mm.
- Material: non-ferrous metal, seawater-resistant
- Type of protection: IP 65
- Seals: oil- and petrol-resistant
- Painting: hammer-finish green RAL 6011
|Casing||Cable glands||Dimension a||Dimension b||Dimension c||Dimension L||Weight [KG]|
|SG 110||5 x PG21||225||244||234||278||9.4|
|SG 111||6 x PG21||285||304||294||338||12.0|
Protective casing SG 150/208SG 150 for absolute encoders
- GEL 152/153 single- or multiturn with parallel or SSI output (cf. Technical Information on encoders, absolute encoder GEL 15X, SSI or parallel interface)
- GEL 152 F for remote programming by PC (cf. Technical Information on encoders, absolute encoder GEL 15X, PC-programmable
- GEL 208 (cf. Technical Information on incremental encoders GEL 207... 219)
Type of protection: IP 68
SG 150/208 with round and square flange
Type Code SG 150
Type Code SG 208