|Publication number||US6643610 B1|
|Application number||US 09/405,196|
|Publication date||4 Nov 2003|
|Filing date||24 Sep 1999|
|Priority date||24 Sep 1999|
|Also published as||CN1376278A, DE60010728D1, DE60010728T2, DE60010728T3, EP1214633A1, EP1214633B1, EP1214633B2, WO2001023972A1|
|Publication number||09405196, 405196, US 6643610 B1, US 6643610B1, US-B1-6643610, US6643610 B1, US6643610B1|
|Inventors||Lowell A. Kleven, Ahmed H. Tewfik|
|Original Assignee||Rosemount Inc.|
|Export Citation||BiBTeX, EndNote, RefMan|
|Patent Citations (39), Non-Patent Citations (38), Referenced by (15), Classifications (23), Legal Events (5)|
|External Links: USPTO, USPTO Assignment, Espacenet|
The invention relates to the field of process measurement and control industry.
The process measurement and control industry employs process variable transmitters to remotely monitor process variables associated with fluids such as flurries, liquids, vapors, gases, chemicals, pulp, petroleum, pharmaceuticals, food and other processing plants. Examples of process variables include pressure, temperature, flow, level, turbidity, concentration, chemical composition, pH and other properties.
Complex mathematical computations are required to determine some process variables. For example, in order to determine flow by measuring differential pressure across an orifice plate, the computation requires the determination of physical properties of the fluid such as fluid density and the gas expansion factor. Calculation of these parameters involves extensive computation which reduces the update time of the transmitter, requires more complex processing equipment and requires increased power.
One technique to reduce the complexity of the calculations is to simply use fixed approximations for some of the parameters. For example, fixed values can be stored in memory rather than calculated using precise formulas. A more accurate technique is the use of straight polynomial curve fitting. In this technique, a process variable is estimated by using a less than complex polynomial rather than perform the precise computation. Such a technique is described in WIPO Publication No. WO 97/04288, filed Jun. 20, 1997 and entitled “TRANSMITTER FOR PROVIDING A SIGNAL INDICATIVE OF FLOW THROUGH A DIFFERENTIAL PRODUCER USING A SIMPLIFIED PROCESS” and U.S. Pat. No. 5,606,513, issued Feb. 25, 1997 and entitled “TRANSMITTER HAVING INPUT FOR RECEIVING A PROCESS VARIABLE FROM A REMOTE SENSOR”. For example, the square root of the reciprocal of the compressibility (Z) of a fluid can be approximated using a “straight” polynomial in the form of:
where P is the absolute pressure of the fluid, T is the absolute temperature and the coefficients Am,n are the coefficients of the interpolating polynomial.
FIGS. 3A and 3B illustrate errors in prior art polynomial curve fitting techniques. FIG. 3A shows the curve fit error for ethylene as a function of pressure for 10 different temperature values. The highest powers of pressure P and inverse temperature (1/T) in the interpolating polynomial are eight and six, respectively. The temperatures and pressures for ethylene are not far from saturation pressures and temperatures. The minimum number (63) of pressure and temperature points was used to determine the 63 coefficients Am,n of the interpolation polynomial set forth in Equation 1, with j=6 and k=8. FIG. 3A illustrates the large error associated with straight polynomial curve fitting, particularly at the pressure extremes. FIG. 3B illustrates an even greater error when 32 bit floating point numbers (with 24 bit mantissa) are used. As illustrated in FIG. 3B, the results are essentially meaningless, with errors exceeding 105 percent. In the example of FIG. 3B, the coefficients were determined with 64 bit numbers and the calculation of the fitting polynomial was performed with 32 bit floating point numbers. Note that 32 bits is the typical number of bits used in floating point math performed in microprocessors. The use of a least squares fitting technique reduces the roughness produced by the loss of significant digits. However, the errors produced with such curve fitting are still relatively large in some situations.
This approximation technique is inaccurate, particularly when the highest power in the polynomial exceeds three or four and the polynomial is formed with multiple independent variables. This inaccuracy can cause inaccuracies in the measured process variable. Typically, the only solution has been to use the exact equations which require powerful computers and high power.
An interpolating polynomial is used to determine a process variable in a transmitter. The interpolating polynomial is “orthogonal” and provides an accurate estimation of the process variable without a significant increase in power consumption. The transmitter includes a sensor configured to couple to a process and having a sensor output related to the process variable. The transmitter also includes a microprocessor coupled to the sensor output and having a process variable output. The process variable output is an orthogonal-polynomial function of the sensor output. A transmitter output is configured to provide an output related to the interpolated process variable.
FIG. 1 is an environmental view of an embodiment of a process transmitter, and in particular a flow transmitter, of the invention.
FIG. 2 is a schematic illustration of a process transmitter.
FIGS. 3A and 3B are prior art graphs of percent error versus pressure for various temperatures for an interpolation of a process variable performed exactly and an interpolation formed using 32 bit floating point numbers, respectively.
FIGS. 4A and 4B are graphs of percent error versus pressure for various temperatures using a Chebychev polynomial interpolation calculated using 64 bit floating point numbers and 32 bit floating point numbers, respectively.
FIGS. 5A and 5B are graphs of percent error versus pressure for various temperatures for a Chebychev polynomial interpolation using a 7×5 matrix using 64 bit floating point numbers and 32 bit floating point numbers, respectively.
FIG. 6 is a block diagram in accordance with one implementation of the invention.
FIGS. 7A and 7B are graphs of percent error versus pressure for various temperatures for a Chebychev polynomial interpolation using 16 bit integer numbers and 32 bit integer numbers, respectively.
FIGS. 8A and 8B are graphs of percent error versus pressure for various temperatures for a Chebychev polynomial interpolation using 16 bit integer numbers and 32 bit integer numbers, respectively, in which the numbers were truncated.
FIG. 1 illustrates the environment of a process transmitter such as flow transmitter 10 of a process measurement or control system 2. Transmitter 10 couples to control room 24 through a two-wire process control loop 14 (shown substantially as voltage source 6 and resistance 8). Transmitter 10 is coupled to a process fluid container such as pipe 12 through pipe fitting or flange 14. Pipe 12 conducts flow of a fluid, such as a gas or liquid, in the direction indicated by arrow 16. The following description sets forth a transmitter such as transmitter 10 for measuring a process variable. The process variable is estimated using a mathematical technique known as polynomial interpolation. A specific type of polynomial, known as an orthogonal-polynomial, is used in the polynomial interpolation. The orthogonal-polynomial provides highly accurate estimations of the process variable without requiring excessive computation. The following description is provided to illustrate and explain aspects of the invention. This description does not limit the scope of the invention and those skilled in the art will recognize that the particular implementation can vary from the specific examples set forth herein.
Transmitter 10 determines flow through pipe 12 by measuring a differential pressure, static pressure and temperature and includes transmitter electronics module 18. Transmitter 10 couples to a resistive temperature device (RTD) carried in temperature sensor housing 24 through electrical conduit 26. Transmitter 10 includes a differential pressure sensor and an absolute pressure sensor. The transmitter 10 provides an output signal indicative of flow of the process fluid flowing through pipe 12 to control room 4 by 4-20 mA two-wire loop 14 preferably formed using twisted pair of conductors through flexible conduit 28. Transmission can be for example, in accordance with the Highway Addressable Remote Transducer (HART®) or Foundations™ Fieldbus standards. Flow is determined using an interpolation polynomial, such as an orthogonal-polynomial in accordance with one aspect of the invention, as will be described later in the specification.
FIG. 2 is a simplified block diagram of a process transmitter 100 which couples to sensor 102 (in some embodiments, element 102 represents multiple sensors) which may be either internal or external to the housing for transmitter 100. The output from sensor 102 is digitized by analog to digital converter 104 and provided to a polynomial interpolator embodied in microprocessor 106. Microprocessor 106 operates at a rate determined by clock 108 and in accordance with instructions stored in memory 110. Memory 110 can also store both permanent and temporary variables. Microprocessor 106 couples to a loop communicator 112 which couples to loop 114.
Transmitter 100 is used to measure a process variable. Sensor 102 is configured to couple to a process, such as process fluid carried in pipe 12 shown in FIG. 1, and provide a sensor output 120 to analog to digital converter 104. Analog to digital converter 104 provides a digitized output 122 to a polynomial interpolator such as microprocessor 106 which provides a process variable output 124 to a transmitter output such as loop communicator 116. The process variable output 124 is a function of an orthogonal-polynomial of sensor output 120. Memory 110 stores the coefficients of the orthogonal-polynomial. The orthogonal-polynomial function is used to approximate the process variable based upon the sensor output using a mathematical method known as interpolation. The measured process variable can be a function of more than one sensor output as described below. Orthogonal-polynomials are known in mathematics and are classes of polynomials pn (x) over a range (a,b) which obey the relationship:
In one aspect, the orthogonal-polynomial function is a particular type of orthogonal polynomial known as a Chebychev polynomial for approximating the process variable through mathematical interpolation. (The Chebychev polynomial is explained below.) In another aspect, an interpolation polynomial is used to approximate the process variable in which the terms of the polynomial are functions of more than one exponential power of the sensor output.
FIG. 2 is provided for illustrative purposes and actual transmitter configurations can vary. For example, the functions performed by microprocessor 106 can be performed by a number of different microprocessors or circuits. The output from sensor 102 can be processed prior to its analog to digital conversion by analog to digital converter 104. Additional compensation steps can be performed using digital circuitry. Many functions can be implemented in either hardware or software, or in their combination. The particular configuration set forth in FIG. 2 should not limit the scope of the invention and those skilled in the art will recognize that the configuration can be modified. For example, sensor 102 can comprise more than one sensor and the interpolated process variable can be a function of more than one sensor output from differing sensors. Typically, each additional sensor output requires an additional term in the interpolation polynomial.
One example of two-wire process control loop carries a current I which has a minimum value of 4 mA and a maximum value of 20 mA. Data can be transmitted in a digital and/or an analog format. Loop communicator 116 is also used by microprocessor 106 to receive data from loop 114. A power module 118 is used to provide power to components in transmitter 100 using power received from loop 114. In some types of transmitters, the transmitter is completely powered with power received from loop 114.
Embodiments of the invention make use of orthogonal-polynomials to form the interpolation equation and solve the resulting system of equations for the coefficients. While many types of orthogonal-polynomials exist, one specific orthogonal-polynomial is a discrete polynomial, known as a discrete orthogonal Chebychev polynomial (hereafter Chebychev polynomial) that provides improved accuracy for process transmitters. In a process transmitter, the independent variables have a finite range and the input and/or calibration data is usually uniformly spaced over the range of the independent variables. Further, the measurements are typically performed with five or six significant digits of accuracy and the required accuracy of the calculated independent variables is in the range of 0.1 to 0.001 percent. Embodiments of the invention can use types of orthogonal-polynomials other than the particular Chebychev polynomial set forth.
FIGS. 4A and 4B are graphs of the error in the curve fit using a Chebychev polynomial in accordance with embodiments of the invention. FIG. 4B illustrates a relatively good curve fit even when 32 bit floating point calculations are used. This level of accuracy can be achieved with prior art straight polynomial curve fitting techniques only if 64 or more bits are used. Although Chebychev polynomial interpolation requires some additional calculations and additions than prior art techniques, the improved accuracy offsets the slight increase in calculation complexity. For example, evaluating a 9×7 straight interpolating polynomial (8TH power in pressure and 6TH power in temperature) requires 62 multiplies and 62 adds. In contrast, a 9×7 Chebychev interpolation requires 88 multiplies and 78 adds. However, this increase in complexity is not particularly significant and only requires a small amount of additional power in view of the improved accuracy in the process variable measurements. Further, the Chebychev interpolation polynomial provides improved accuracy, even when a smaller number of bits are used for the calculation. The curve fit improves sufficiently using the Chebychev polynomial that the number of terms in the fitting polynomial can be reduced while maintaining acceptable accuracy. FIG. 5A shows the accuracy of a 7×5 exact Chebychev interpolation and FIG. 5B illustrates errors when 32 bit floating point numbers are used. The results compare favorably to a 9×7 least square curve fit. Further, in the 7×5 Chebychev polynomial curve fit the number of multiplies is reduced to 52 and number of additions is reduced to 46, which is less than those required by a 9×7 straight polynomial interpolation.
An nth order Chebychev polynomial can be denoted as Tn (x). The highest power of x in Tn (x) is n. Discrete Chebychev polynomials are orthogonal over a discrete set of integers, 0≦k≦N−1, m,n≦N−1. Specifically, orthogonal-polynomials are denoted as:
One specific Chebychev polynomial obeys the recursive formula:
in which, T0(x)=1, T1(x)=2X−(N−1), n is the order of the Chebychev polynomial and N is the number of independent pressure and temperature readings that are used to determine the coefficients of the interpolating linear combination of Chebychev polynomials. For example, if N=7 Tn(x) can be expanded as:
T 4(x)=70x 4−840x 3+3110 x 2−3540x=360 (9)
Note that n≦N−1 or the highest order of the Chebychev polynomial must be less than the number of independent pressure or temperature points. As illustrated in Equations 6-11, in one aspect polynomial terms are used which include more than one exponential power of the sensor output x.
In one example, an orthogonal-polynomial approximation is used to obtain density (ρ) using the gas law ρ=P/ZRT. R is the gas constant of the fluid being used and ρ is the density. To perform a curve fitting to 1/Z where Z is the compressibility, (1/Z is used to avoid divisions in the microprocessor), Equation 12 is used:
where P is pressure, P is the minimum pressure, S is 1/T, T is temperature, Smin is the minimum value of 1/T, NPE is the number of coefficients for pressure, NTE is the number of coefficients for temperature, xp=(P−Pmin)/ΔP, xs=(S−Smin)/ΔS,S=1/T, NPE−1 is the largest power of pressure or order of the Chebychev polynomial used for pressure, NTE−1 is the largest power of temperature or order of the Chebychev polynomial used for temperature, ΔP=(Pmax−Pmin)/(NPt−1), ΔS=(1/Tmin−1/Tmax)/(NTt−1), Am,n are the coefficients of the Chebychev interpolation formula and NPt and NTt are the numbers of pressure points and temperature points, respectively, used in the curve fit. The range of xp is from 0 to NPt−1 and xs is from 0 to NTt−1.
The coefficients Am,n are solved by forming Chebychev matrices corresponding to the pressure (P) and temperature (S=1/T) variables. These matrices are shown below:
The columns of the matrices Q1 and Q2 are normalized by dividing by the square root of the sum of the squares of all entries in each column. For example, the normalization factor for the nth column of Q1 is:
The normalization factors for the nth column of Q2 is:
The normalization factors are determined for the case when:
When P and S are uniformly spaced, and xpm and xsn are integers as shown above, this normalization turns the matrices Q1 and Q2 into orthogonal matrices with a unit condition number. Specifically, this provides Q1 T•Q1=INTE and Q2 T•Q2=INTE, where Ik denotes a k×k identify matrix.
Equation 12 can be rewritten in matrix form as:
In the above equation, Q2 T denotes the transpose of Q2, A is an NPE×NTE matrix whose (m,n)th entry is coefficient Am,n and Z is an NPt×NTt matrix whose (m,n)th entry is 1/Z(xpm, xsn). When xpm and xsn are integers, Equation 19 is easily solved as:
Equation 20 holds and is well-behave numerically because both Q1 and Q2 are orthogonal matrices with unit condition numbers.
One property of the above solution is that the magnitude of the elements of matrix A can be inspected to determine the relative importance of each term in the Chebychev interpolation formula. Small terms can be dropped from the calculation. Table 1 below shows the relative magnitudes of the entries of the A matrix that corresponds to the 9×7 Chebychev interpolation that we discussed in the previous section. The magnitudes of the entries of A are displayed as percentage of the magnitude of the largest coefficient. Columns correspond to temperatures and rows to pressure. As can be seen, the use 9×7 is probably not justified by the size of the coefficients.
In order to have a unit condition number for matrices Q1 and Q2 and maintain orthogonality, the values for xp and xs have to be uniformly spaced over their range. However, this requirement does not have to be strictly enforced to make use of Chebychev polynomials. When xpm and xsn are not integers, Equation 19 can be solved directly or in a least squares sense as:
Here (Q1 TQ1)−1 denotes the inverse of (Q1 TQ1) and Q2 −T that of Q2 T. Equation 21 is reasonably well-behaved numerically. In experiments, the condition numbers of both Q1 and Q2 remained less than 450. While the above matrix Equation is formidable, most numerical computer math packages have special routines to solve these equations. The microprocessor 106 in transmitter 100 does not have to solve these equations. The equations are calculated off line and only the coefficient (stored in memory 110) are used in the microprocessor 106 as explained below. These equations can also be expanded and solved by Gaussian elimination techniques.
The normalization factors, norp(m) and nort(n), used to normalize Q1 and Q2 are always determined from uniformly spaced xpm and xsn, and not from the actual xpm and xsn. Several tests with the normalization factors determined from the actual non-uniform spacing of the data did not produce any substantial difference in accuracy.
The sensitivity of the Chebychev interpolation approach to non-uniformly spaced pressure and inverse temperature points was tested. The curve fit involved 7 temperatures and 9 pressure for ethylene. The curve fit errors of the uniformly spaced data are shown in FIGS. 4A and 4B. Each temperature and pressure point used in the curve fit was perturbed from its uniform position by a random amount uniformly distributed over the range±½ the uniform step size. Theoretically, this perturbation approach could lead to two identical pressure or temperature points. However, this did not happen during the test. Instead, the condition numbers for Q1 varied from 2 to 440 and those of Q2 from 1.2 to 65. The maximum error of the curve fit varied from 0.00034% to 0.0014%. The maximum fit error in FIGS. 4A and 4B was 0.00075%. This does not represent a significant change in the curve fit error.
A test was performed with an overdetermined system of equations by increasing the number of temperature and pressure points by one. In this case, the maximum condition numbers decreased to 34 and 19 for Q1 and Q2, respectively. The maximum curve fit error changed to 0.0012% as compared to a maximum fit error of 0.0006% for the uniform case. As more points are used in the curve fit, the condition numbers approach one and the maximum fit error approaches 0.00018% under the same test conditions as above. That maximum fit error compares favorably with a maximum fit error of 0.00015% when a large number of uniformly spaced pressure and inverse temperature points are used.
The numerical complexity of the Chebychev interpolation approach can be reduced by renormalizing the Chebychev polynomials. Tp m(xp) and Ts m(xs) are the normalized pressure and temperature Chebychev terms, respectively. These normalized polynomials are equal to the regular Chebychev polynomials (Equation 4) divided by a constant that depends on the order of the polynomial and the number of points used in the interpolation. They satisfy the following equations:
In the above Equations:
T 1 P(xp)=Cp(1)(P−P min)−1 (24)
Note that S=1/T and Smin=1/Tmax where T is the absolute temperature. The values of pressure and temperature must be between the maximum and minimum pressure and temperature points used in the curve fitting.
To use the normalized Chebychev polynomials, the coefficients Am,n must be normalized appropriately. Bm,n denotes the normalized interpolation coefficients. The coefficients Bm,n are related to the coefficients Am,n through the formula:
The normalization factors Cp(m), Ct(n), norpt(m) and nortt(n) are evaluated as follows. The normalization factors, norp(m) and nort(n), used to form matrices Q1 and Q2 in the solution for Am,n are required to calculate these parameters. Initialize the normalization coefficients with the following relations:
Next, perform the following recursion with m−2 to m−NPE−1 for the pressure points:
and the following recursion with n=2 to n=NTE−1 for the temperature points:
Finally, compute the independent variable, 1/Z in this case, in the transmitter using the Equation:
along with Equations 22, 23, 24 and 25. The zero order Chebychev polynomial is not calculated since its value is 1.
In summary, given the coefficient array Bm,n, the vectors Cp(m), Ct(n), Pmin and Smin, the calculation procedure for the microprocessor 106 in the transmitter 100 using the method 150 shown in FIG. 6. The method starts at start block 152. At block 154, microprocessor 106 retrieves the Chebychev polynomial coefficients from memory 110. At 156 sensor outputs 120, such as pressure and temperature, are obtained from the sensors and the first order Chebychev polynomial is calculated using Equations 24 and 25 at block 158. At 160, the remaining Chebychev polynomials are calculated using Equations 22 and 23. The process variable 1/Z is approximated using Equation 34 by microprocessor 106 at block 162. At block 164, the approximated process variable is output, for example, using loop communicator 116. At 166, control is returned to start block 152 and the interpolation process repeats.
To illustrate the required multiplications and additions, equation (34) is expanded below:
Note that the T0(x) term has been dropped since its value is one. This equation requires NTE(NPE−1)+NTE−1 multiplies and adds. The Chebychev recursions (Equations 22, 23, 24, and 25) require 2(NPE+NTE)−6 multiplies and NPE+NTE adds. Therefore, the total complexity of the Chebychev interpolation approach is NPE×NTE+2(NPE+NTE)−7 multiples and NPE×NTE+NPE+NTE−1 adds. This is to be compared to the direct polynomial interpolation which requires NPE×NTE−1 multiplies and adds. This is an increase in the multiples and adds, but the benefits are great.
All of the above equations can be evaluated in integer math in microprocessor 106 provided that an additional normalization process is performed. If all the operations involve numbers in the range of ±1, then the results will also be in the same range. In this case, all numbers can be rescaled to 2n−1, where n is the number of 2's complement bits used in the calculation. The recursion relations given in Equations 22 and 23 produce values that fall in the range −1 to +1. The Cp and Ct normalization coefficients are also less than one with the exception of Cp(1) and Ct(1) that convert the temperature and pressure to T1(x). Therefore, if the input numbers representing pressure and temperature fit the integer format being used, Equation 24 and 25 can also be implemented in integer format.
To evaluate Equation 34 using integer math, two normalizations are performed. The first normalization scales 1/Z to have a maximum value of one. The second normalization scales the values of the coefficients Bm,n so that the summations in Equation 34 do not produce a value greater than one.
To normalize 1/Z, divide all the entries 1/Z(xpm,xsn) of matrix Z by a constant Kf, where Kf is the maximum value of all the entries 1/Z(xpm, xsn) of matrix Z.
To normalize the coefficients Bm,n, solve for matrix A and produce matrix B whose entries are the coefficients Bm,n. Matrix B is then reduced by dividing through by nor, where nor is the sum of the absolute value of all the elements in matrix B. C denotes the resulting matrix. The entries Cm,n of C are related to the coefficients Bm,n through Cm,n=Bm,n/nor. This normalization guarantees that no summation will exceed 1. This is the most conservative normalization approach and guarantees that no summations will exceed 1.
Another normalization approach consists of sweeping the xp and xs values over their range and finding the maximum value of the resulting Tn(xs) and Tm(xp). This maximum value is used in conjunction with the appropriate Bm,n coefficient to determine the maximum value that the sum can achieve. This value is then used as the normalization value nor. This normalization approach has the advantage of making better use of the available number range.
After the microprocessor 106 has calculated the normalized value of 1/Z, the correct value can be obtained by multiplying by Kf and nor. Specifically, 1/Z is computed as:
The constants Kf and nor are combined with some other constant such as the gas constant, R. The range of the input pressures and temperatures must be limited to slightly less then the curve, fit range so that round off errors do not cause the variables to exceed the maximum and minimum curve fit pressures and temperatures. Less than 1% reduction in range is necessary to achieve this condition.
FIGS. 7A and 7B show the curve fit errors using 16 bit and 32 bit integer math, respectively. The 16 bit calculation shows that numerical errors are present. However, the magnitude of these errors is less than 3 or 4 least significant bits. For most transmitters, this accuracy is sufficient. These error curves were produced using rounding for all multiplications and converting numbers to 16 bit integers. If truncation is used, the errors are larger and produce systematic errors. With truncation, the errors approach 0.08% for 16 bit integers. FIGS. 8A and 8B show the results of truncation for 16 bit and 24 bit integers, respectively. The errors due to truncation just begin to appear when 24 bit integers are used.
Although the present invention has been described with reference to specific embodiments, workers skilled in the art will recognize that changes may be made in form and detail without departing from the spirit and scope of the invention. For example, the polynomial interpolator can be implemented in a microprocessor circuitry. In some aspects, orthogonal-polynomials other than the specific Chebychev polynomial disclosed herein are used. The polynomial can be used to estimate a process variable using any number of sensor outputs. For example, pressure can be corrected based upon a temperature reading and flow can be calculated based upon differential pressure, static pressure and temperature. The orthogonal-polynomial provides an accurate interpolation of a process variable without requiring highly complex calculations which have large electrical power requirements.
|Cited Patent||Filing date||Publication date||Applicant||Title|
|US3564912||28 Oct 1968||23 Feb 1971||Westinghouse Electric Corp||Fluid flow measurement system|
|US3701280||18 Mar 1970||31 Oct 1972||Daniel Ind Inc||Method and apparatus for determining the supercompressibility factor of natural gas|
|US4103551||31 Jan 1977||1 Aug 1978||Panametrics, Inc.||Ultrasonic measuring system for differing flow conditions|
|US4238825||2 Oct 1978||9 Dec 1980||Dresser Industries, Inc.||Equivalent standard volume correction systems for gas meters|
|US4249164||14 May 1979||3 Feb 1981||Tivy Vincent V||Flow meter|
|US4403296||18 Dec 1980||6 Sep 1983||Electromedics, Inc.||Measuring and determination device for calculating an output determination based on a mathematical relationship between multiple different input responsive transducers|
|US4414634||17 Jul 1981||8 Nov 1983||The Scott & Fetzer Company||Fluid flow totalizer|
|US4419898||15 Oct 1981||13 Dec 1983||Sarasota Automation Limited||Method and apparatus for determining the mass flow of a fluid|
|US4437164||5 Mar 1981||13 Mar 1984||Bristol Babcock Inc.||Ridge circuit compensation for environmental effects|
|US4446730||26 Apr 1982||8 May 1984||Quintex Research International, Inc.||Specific gravity independent gauging of liquid filled tanks|
|US4485673||13 May 1981||4 Dec 1984||Drexelbrook Engineering Company||Two-wire level measuring instrument|
|US4528855||2 Jul 1984||16 Jul 1985||Itt Corporation||Integral differential and static pressure transducer|
|US4562744||4 May 1984||7 Jan 1986||Precision Measurement, Inc.||Method and apparatus for measuring the flowrate of compressible fluids|
|US4598381||24 Mar 1983||1 Jul 1986||Rosemount Inc.||Pressure compensated differential pressure sensor and method|
|US4602344||25 Oct 1984||22 Jul 1986||Air Products And Chemicals, Inc.||Method and system for measurement of liquid level in a tank|
|US4677841||6 Mar 1986||7 Jul 1987||Precision Measurement, Inc.||Method and apparatus for measuring the relative density of gases|
|US4796651||30 Mar 1988||10 Jan 1989||LeRoy D. Ginn||Variable gas volume flow measuring and control methods and apparatus|
|US4799169||24 Feb 1988||17 Jan 1989||Mark Industries, Inc.||Gas well flow instrumentation|
|US4825704||24 Aug 1987||2 May 1989||Yamatake-Honeywell Co., Ltd.||Fluid flow speed measuring apparatus|
|US4864462||8 Nov 1988||5 Sep 1989||Sri International||Capacitive gas detector having a solid rare earth fluoride dielectric|
|US4881412||4 Aug 1988||21 Nov 1989||Ronald Northedge||Flow meters|
|US4958938||5 Jun 1989||25 Sep 1990||Rosemount Inc.||Temperature transmitter with integral secondary seal|
|US4986127||5 Apr 1989||22 Jan 1991||Hitachi, Ltd.||Multi-functional sensor|
|US5035140||3 Nov 1988||30 Jul 1991||The Boeing Company||Self cleaning liquid level detector|
|US5058437||22 Sep 1989||22 Oct 1991||Gaz De France||Determining the quantity yield of a compressible fluid flowing through a pressure reducing valve|
|US5495769||22 May 1995||5 Mar 1996||Rosemount Inc.||Multivariable transmitter|
|US5606513||9 Jun 1994||25 Feb 1997||Rosemount Inc.||Transmitter having input for receiving a process variable from a remote sensor|
|US5772323||26 Oct 1994||30 Jun 1998||Felice; Ralph A.||Temperature determining device and process|
|US5857777 *||25 Sep 1996||12 Jan 1999||Claud S. Gordon Company||Smart temperature sensing device|
|US5949695 *||10 Jan 1997||7 Sep 1999||Harris Corporation||Interpolator using a plurality of polynomial equations and associated methods|
|US6182019 *||20 Jun 1997||30 Jan 2001||Rosemount Inc.||Transmitter for providing a signal indicative of flow through a differential producer using a simplified process|
|EP0214801A1||22 Aug 1986||18 Mar 1987||Parmade Instruments C.C.||A method of monitoring the liquid contents of a container vessel, monitoring apparatus for use in such method, and an installation including such apparatus|
|EP0223300A2||11 Nov 1986||27 May 1987||EMS Holland BV||Gas meter|
|WO1988001417A1||12 Aug 1987||25 Feb 1988||Rosemount Inc.||Analog transducer circuit with digital control|
|WO1989002578A1||16 Sep 1988||23 Mar 1989||Square D Company||Modular switch device|
|WO1989004089A1||23 Sep 1988||5 May 1989||Rosemount Inc.||Transmitter with internal serial bus|
|WO1990015975A1||14 Jun 1990||27 Dec 1990||Rosemount Inc.||Extended measurement capability transmitter having shared overpressure protection means|
|WO1991018266A1||16 May 1991||28 Nov 1991||A.G. (Patents) Limited||Fluid pressure operated volume measurement with level calibration means|
|WO1997004288A1||11 Jul 1996||6 Feb 1997||Rosemount Inc.||Transmitter for providing a signal indicative of flow through a differential transducer using a simplified process|
|1||"A Systems Approach", by Dr. C. Ikoku, Natural Gas Engineering, 2 pgs. (undated).|
|2||"Compressibility Factors of Natural Gas and Other Related Hydrocarbon Gases", AGA Transmission Measurement Committee Report No. 8, American Petroleum Institute MPMS Chapter 14.2, Gas Research Institute, Catalog No. XQ9212, Second Edition, Nov. 1992, 2nd Printing Jul. 1994.|
|3||"Development of an Integrated EFM Device for Orifice Meter Custody Transfer Applications," S.D. Nieberle et al., American Gas Association Distribution/Transmission Conference & Exhibit, May 10, 1993, 5 pages.|
|4||"Digital Computers For Gas Measuring Systems", by Robert D. Goodenough, 8131 Advances in Instrumentation, vol. 31, No. 4 (1976), pp. 1-4 (no month).|
|5||"Flow Measurement," Handbook of Fluid Dynamics, V. Streeter, Editor-in-chief, published by McGraw-Hill Book Company, Inc. 1961, pp. 14-4 to 14-15 (No month).|
|6||"Hydrostatic Tank Gauging-Technology Whose Time Has Come," J. Berto, Rosemount Measurement Division Product Feature, undated reprint from INTECH, 2 pages.|
|7||"Low Cost Electronic Flow Measurement System," Tech Profile, May 1993, Gas Research Institute, Chicago, IL, 2 pages.|
|8||"Methods for Volume Measurement Using Tank-Gauging Devices Can Be Error Prone," F. Berto, The Advantages of Hydrostatic Tank Gauging Systems, undated reprint from Oil & Gas Journal, 7 pages.|
|9||"Model 3095 Flow Transmitter," Preliminary Product Information, September 1993, Rosemount Inc., Eden Prairie, MN 55344, 11 pages.|
|10||"Numerical Recipes in C", by W. Press et al., Cambridge University Press, 1990, pp. 51-59 and 158-162.|
|11||"Numerical Recipes in Fortran", by W. Press et al., Cambridge University Press, 1992, pp. 650-651 and 664-665 (no month).|
|12||"Numerical Recipes in Fortran", by W. Press et al., Cambridge University Press, 1992, pp. 99-122.|
|13||"Orifice Metering Of Natural Gas and Other Related Hydrocarbon Fluid", Part 1, General Equations and Uncertainty Guidelines, American Gas Association, Report No. 3, American Petroleum Institute, API, 14.3, Gas Processors Association, GPA 8185-90, Third Edition, Oct. 1990, A.G.A. Catalog No. XQ9017.|
|14||"Orifice Metering Of Natural Gas and Other Related Hydrocarbon Fluid", Part 2, Specification and Installation Requirements, American Gas Association, Report No. 3, American Petroleum Institute, API 14.3, Gas Processors Association, GPA 8185-90, Third Edition, Feb. 1991, A.G.A. Catalog No. XQ9104.|
|15||"Orifice Metering Of Natural Gas and Other Related Hydrocarbon Fluid", Part 3, Natural Gas Applications, American Gas Association, Report No. 3, American Petroleum Institute, API 14.3, Gas Processors Association, GPA 8185-92, Third Edition, Aug. 1992, A.G.A. Catalog No. XQ9210.|
|16||"Orifice Metering Of Natural Gas and Other Related Hydrocarbon Fluid", Part 4, Background, Development, Implementation Procedure, and Subroutine Documentation for Empirical Flange-Tapped Discharge Coefficient Equation, American Gas Association, Report No. 3, American Petroleum Institute, API 14.3, Gas Processors Association, GPA 8185-92, Third Edition, Oct. 1992, 2nd Printing Aug. 1995, A.G.A. Catalog No. XQ9211.|
|17||"Precise Computerized In-Line Compressible Flow Metering," Flow-Its Measurement and Control in Science and Industry, vol. 1, Part 2, Edited by R. Wendt, Jr., Published by American Institute of Physics et al, (undated) pp. 539-540.|
|18||"Pressure Sensors Gauge Tank Level and Fluid Density," Rosemount Measurement Division Product Feature, undated reprint from Prepared Foods (Copyrighted 1991 by Gorman Publishing Company), 2 pages (no month).|
|19||"Signal Transmission Put On A Pedestal", Control and Instrumentation, Sep., 1976, vol. 6, No. 8, pp. 28-29.|
|20||"Single Chip Senses Pressure and Temperature," Machine Design, 64 (1992) May 21, No. 10, 1 page.|
|21||"Smart Transmitters Tear Up The Market," C. Polsonetti, INTECH, Jul. 1993, pp. 42-45.|
|22||"The Digitisation of Field Instruments" W. Van Der Bijl, Journal A, vol. 32, No. 3, 1991, pp. 62-63 (No month).|
|23||"Hydrostatic Tank Gauging—Technology Whose Time Has Come," J. Berto, Rosemount Measurement Division Product Feature, undated reprint from INTECH, 2 pages.|
|24||"Precise Computerized In-Line Compressible Flow Metering," Flow—Its Measurement and Control in Science and Industry, vol. 1, Part 2, Edited by R. Wendt, Jr., Published by American Institute of Physics et al, (undated) pp. 539-540.|
|25||Advertisement, AccuRate Model 3308 Integral Smart DP/P/T Transmitter, (undated) Bristol Babcock, Inc., Watertown, CT 06795, 1 page, (no date).|
|26||Advertisement, Model 3508 DP Transmitter, Control Engineering, Dec. 1992, Bristol Babcock, Inc., Watertown, CT 06795, 1 page.|
|27||Brochure: "Reduce Unaccounted-For Natural Gas with High-Accuracy Pressure Transmitters," Rosemount Inc. Measurement Division, Eden Prairie, Minnesota, ADS 3073, 5/91, pp. 1-4.|
|28||Ghosh D. et al., "Linearization of Transducers Through a Generalized Software Technique," Measurement Science and Technology, GB, IOP Publishing, Bristol, vol. 2, 1991, pp. 102-105 (no month).|
|29||P. Mahana et al., "Transducer Output Signal Processing Using an Eight-Bit Microcomputer," IEEE Transactions on Instrumentation and Measurement, No. 2, Jun. 1986 (1986-06), pp. 182-186.|
|30||Product Data Sheet 4638, "Model 3001CL Flush-Mount Hydrostatic Pressure Transmitter," July 1992, Rosemount Inc., Eden Prairie, MN 55344, 3 pages.|
|31||Product Data Sheet 4640, "Model 3201 Hydrostatic Interface Unit," Mar. 1992, Rosemount Inc., Eden Prairie, MN 55344, 8 pages.|
|32||Product Data Sheet PDS 4015, "Model 3095FT Flow Transmitters", by Rosemount Inc. Eden Prairie, MN 55344, Feb. 1994, pp. 1-15.|
|33||Specification Summary, "AccuRate Advanced Gas Flow Computer, Model GFC 3308," (undated) Bristol Babcock, Inc., Watertown, CT, 06795, 12 pages.|
|34||Specification Summary, "TELETRANS(TM) 3508-10A Smart Differential Pressure Transmitter," (undated) Bristol Babcock, Inc., Watertown, CT, 06795, 10 pages.|
|35||Specification Summary, "TELETRANS(TM) 3508-30A Smart Differential Pressure Transmitter," (undated) Bristol Babcock, Inc., Watertown, CT, 06795, 11 pages.|
|36||Specification Summary, "TELETRANS™ 3508-10A Smart Differential Pressure Transmitter," (undated) Bristol Babcock, Inc., Watertown, CT, 06795, 10 pages.|
|37||Specification Summary, "TELETRANS™ 3508-30A Smart Differential Pressure Transmitter," (undated) Bristol Babcock, Inc., Watertown, CT, 06795, 11 pages.|
|38||Technical Information Bulletin, "Liquid Level Transmitter Model DB40RL Sanitary Sensor deltapilot," Endress+Hauser, Greenwood, Indiana, 9/92, pp. 1-8.|
|Citing Patent||Filing date||Publication date||Applicant||Title|
|US7209837||4 Jun 2004||24 Apr 2007||Invensys Systems, Inc.||Systems and methods for determining compensated conductivities|
|US7768530 *||25 Jan 2007||3 Aug 2010||Fisher-Rosemount Systems, Inc.||Verification of process variable transmitter|
|US8209039||26 Jun 2012||Rosemount Inc.||Process control system having on-line and off-line test calculation for industrial process transmitters|
|US8578783||26 Sep 2011||12 Nov 2013||Rosemount Inc.||Process fluid pressure transmitter with separated sensor and sensor electronics|
|US8655604||27 Oct 2009||18 Feb 2014||Rosemount Inc.||Multivariable process fluid flow device with fast response flow calculation|
|US8849589||22 May 2009||30 Sep 2014||Rosemount Inc.||Multivariable process fluid flow device with energy flow calculation|
|US20040075685 *||10 Feb 2003||22 Apr 2004||Fuji Xerox Co., Ltd.||Usability evaluation support apparatus and method|
|US20080180276 *||25 Jan 2007||31 Jul 2008||Fisher-Rosemount Systems, Inc.||Verification of process variable transmitter|
|US20090292484 *||22 May 2009||26 Nov 2009||Wiklund David E||Multivariable process fluid flow device with energy flow calculation|
|US20100082122 *||1 Apr 2010||Rosemount Inc.||Process control system having on-line and off-line test calculation for industrial process transmitters|
|US20100106433 *||27 Oct 2009||29 Apr 2010||Kleven Lowell A||Multivariable process fluid flow device with fast response flow calculation|
|CN101600944B||18 Jan 2008||20 Jul 2011||费希尔-罗斯蒙德系统公司||Verification of process variable transmitter|
|EP2347224A1 *||27 Oct 2009||27 Jul 2011||Rosemount, Inc.||Multivariable fluid flow measurement device with fast response flow calculation|
|WO2013048598A1 *||14 Jun 2012||4 Apr 2013||Rosemount, Inc.||Process fluid pressure transmitter with separated sensor and sensor electronics|
|WO2015131025A1 *||27 Feb 2015||3 Sep 2015||Phadke Associates, Inc.||Method and apparatus for rapid approximation of system model|
|U.S. Classification||702/183, 702/140, 702/138, 702/45|
|International Classification||G05B13/04, G06F17/10, G06F17/17, G08C15/00, G08C13/00, G05B23/02, G01D21/00, G08C19/02, G05B19/042|
|Cooperative Classification||G05B2219/31126, G05B2219/37401, G01F1/42, G01F25/0007, G05B2219/34142, G05B19/0423, G05B2219/34084|
|European Classification||G05B19/042N, G01F1/42, G01F25/00A|
|24 Sep 1999||AS||Assignment|
Owner name: ROSEMOUNT INC., MINNESOTA
Free format text: ASSIGNMENT OF ASSIGNORS INTEREST;ASSIGNORS:KLEVEN, LOWELL A.;TEWFIK, AHMED H.;REEL/FRAME:010273/0797
Effective date: 19990923
|28 Sep 2004||CC||Certificate of correction|
|5 Apr 2007||FPAY||Fee payment|
Year of fee payment: 4
|2 May 2011||FPAY||Fee payment|
Year of fee payment: 8
|4 May 2015||FPAY||Fee payment|
Year of fee payment: 12