If you are looking for chemical, biology, pharmaceutical, and food 1㎡ vacuum freeze dryer from China 1㎡ vacuum freeze dryer, 1㎡ vacuum lyophilizer manufacturers, suppliers and facoty, or checking 1㎡ vacuum freeze dryer customization and desing with professional supplier in China, please feel free to contact Toption Instrument.
TPV-100F Ordinary Type Vacuum Freeze Dryer
● The process of prefreezing and drying finishes in situ. The freeze dryer is equipped with observation windows and the drying process is automatic;
● The Temperature error is less than or equal 1℃. The drying effect is uniform;
● The partition-board temperature is adjustable and controlled;
● Stored 32programs, each program can be set up 36 segments, correct program data in computing, memorize the drying curve;
● The drying chamber and cold trap is separated so the water-capture capability is higher and the drying time is shorter;
● Touching-screen operation, PID controlling, displaying drying curve;
● The sample tray is square so it’s easy to clean and operate;
●Configuration of the inflation valve, rechargeable dry inert gas;
● The drying chamber adopts transparent organic glass door the changing process of material is clear;
● The test device is optional.
Note: TOPT-100F control system is PLC.
1. freeze-drying area：1.0㎡
2. layers of partition board：6+1 layers
3. temperature scope of partition board：-50℃～70℃
4. partition board temperature error: ±1(ambulance)
5. partition board space：70mm
6. size of partition board：360*480*15
7. cold trap temperature：≤-70℃（empty）
8. water-capture capability：15kg/24h
10. defrost style: electric defrost
11. vacuum degree：≤10Pa
13. weight: 680Kg