|Publication number||US5672849 A|
|Application number||US 08/221,748|
|Publication date||30 Sep 1997|
|Filing date||31 Mar 1994|
|Priority date||31 Mar 1994|
|Publication number||08221748, 221748, US 5672849 A, US 5672849A, US-A-5672849, US5672849 A, US5672849A|
|Inventors||L. Dale Foster, Ryan Anthony Reeder|
|Original Assignee||Hill-Rom Company, Inc.|
|Export Citation||BiBTeX, EndNote, RefMan|
|Patent Citations (14), Referenced by (43), Classifications (27), Legal Events (6)|
|External Links: USPTO, USPTO Assignment, Espacenet|
This invention relates generally to a patient care apparatus, and more particularly to a patient weigh scale for weighing a bedridden patient.
Boom type sling scales have heretofore been employed for weighing of bedridden patients. Such a boom type scale has a base, an upright support pole connected to the base, and a cantilever boom connected to the upper end of the support pole. One or more cantilever beam type load cells are supported from the boom and are operatively connected to a sling for receiving a patient therein. A piston and cylinder is connected between the upright support pole and the cantilever boom. In use, the sling is placed adjacent a patient on a hospital bed, and the patient is rolled into the sling (known as "log rolling"). The sling is then operatively connected to the load cells, and the piston and cylinder is actuated to raise the boom and hence the patient up off of the hospital bed. The patient's weight is then taken. Such boom type sling scales are shown in U.S. Pat. Nos. 4,420,052; 4,482,783; and 5,033,563.
Such traditional boom type scales suffer from a number of disadvantages. These traditional boom type scales, with boom affixed generally perpendicular to the upright support pole via the fixed piston and cylinder arrangement, cannot be compactly configured for storage purposes.
In addition, such prior art boom type scales, which include casters for rolling movement of the scales, allow undesirable relative motion between the scale and the bed when weighing a patient.
Further, the so-called "log rolling" required to place a patient into a traditional sling scale consumes time and effort of care providers as well as results in somewhat rough treatment of the patient.
Still further, when a traditional sling scale is employed, the patient's heart rate tends to rise as the patient becomes squeezed in the sling and is simultaneously lifted upwardly. And by raising the patient relative to the floor, IV lines from floor supported infusion equipment can become undesirably taught leading to IV line pullout with associated delitereous effects on sutures, etc.
It has therefore been an objective of the present invention to provide a patient weigh scale which can be compactly configured for storage.
It has been another objective of the present invention to provide a patient weigh scale which eliminates relative motion between the scale and hospital bed during weighing of the patient.
It has yet been another objective of the present invention to provide a patient weigh scale and method of weighing a patient which avoids log rolling of the patient into a traditional sling scale.
It has been still another objective of the present invention to avoid IV line pullout and associated adverse effects on sutures when weighing a patient.
The stated objectives are attained by a patient weigh scale of the present invention which comprises a base, an upright support pole mounted to the base, a cantilever boom pivotally mounted to an upper end of the upright support pole, a transducer suspended from the cantilever boom for sensing the weight of a load applied to the scale, and a bracing arm pivotally mounted on one end to one of the upright support pole and cantilever boom and removably securably on the other end to the other of the pole and boom. The arm, when secured to the other of the pole and boom, fixes the boom generally perpendicular to the pole for weighing a patient, and when removed from the other of the upright support pole and boom allows the boom to fold downwardly generally parallel the pole into a compact storage position.
Preferably the bracing arm is pivotally mounted on one end to the pole, and includes a through hole on the other end alignable with a similar through hole in the boom when the boom is generally perpendicular to the pole. A pull pin is inserted through both holes for fixing the boom generally perpendicular to the pole, and is removed for configuring the scale into the compact storage configuration.
The present invention further provides a weigh scale comprising a base, an upright support pole mounted to the base, a cantilever boom mounted on one end to an upper end of the upright support pole, a transducer suspended from the cantilever boom for sensing the weight of a load applied to the scale, and a latch mounted to one of the base and pole for removably securing the scale to a hospital bed while weighing a patient. The latch secures the weigh scale to the bed and prevents relative movement between the scale and the bed during patient weighing.
The latch preferably comprises a latch block, a latch finger pivotally mounted to the block for pivoting about a pivot axis, the latch finger projecting forwardly of the pivot axis, and a foot actuatable pedal fixedly secured to the latch finger and projecting rearwardly of the pivot axis. The pedal is operable to pivot the latch finger upwardly out of engagement with a hospital bed base mounted latch bar when depressed.
The latch block is preferably rotatably mounted to the pole, such that when the scale is latched to the hospital bed, the scale may be rotated about a generally vertical axis relative to the bed to align the scale with the bed.
The scale further includes a cross member pivotally mounted to the other end of the boom for pivoting about a generally vertical axis relative to the boom, with the transducer being suspended from the cross member. When the scale is latched to the hospital bed, the cross member and transducer may be rotated about a generally vertical axis relative to the boom to align the transducer with the bed.
In the preferred configuration of the patient weigh scale of the present invention, a pair of transducers are mounted on either end of the cross member for sensing the weight of a load applied to the scale. The transducers are preferably cantilever beam type load cells oriented transversely to the cross member and having fixed ends mounted to the cross member and free ends laterally spaced from the fixed ends. An H-frame is suspended from the load cells, with the free ends of the load cells being attached to the arms of the H-frame, and the ends of the H-frame arms being for connection to lateral sides of a patient support.
The present invention is further predicated upon a method of weighing a patient atop a patient support, the patient support being mounted on a raisable and lowerable frame on the hospital bed. The method comprises providing a patient weigh scale having a base, an upright support pole mounted to the base, a cantilever boom mounted to an upper end of the upright support pole, a transducer suspended from the cantilever boom for sensing the weight of a load applied to the scale and a display; connecting the transducer to lateral sides of the patient support; lowering the frame out from under the patient support; and noting the patient weight on the display.
The method may further include the step of removably securing the scale to the hospital bed prior to connecting the transducer to the support to prevent relative movement between the scale and the bed during patient weighing.
The method may still further include the step of rotating the scale about a generally vertical axis relative to the bed to align the scale with the bed after removably securing the scale to the bed and prior to connecting the transducer to the support.
The method may yet further include the step of rotating the transducer about a generally vertical axis relative to the boom to align the transducer with the bed after removably securing the scale to the bed and rotating the scale relative to the bed and prior to connecting the transducer to the support.
One advantage of the present invention is that a patient weigh scale is provided which can be compactly configured for storage purposes.
Another advantage of the present invention is that a patient weigh scale is provided which conveniently docks to a hospital bed thereby eliminating relative movement between the bed and the scale during patient weighing.
Yet another advantage of the present invention is that a method of patient weighing is provided which avoids log rolling a patient into a traditional sling scale.
Still another advantage of the present invention is that a patient is not lifted from a mattress with a sling with the normally attendant adverse consequences on IV lines, sutures and heart rate. Rather, the bed frame is simply lowered from underneath the rigid patient support and the patient's weight is taken. The IV lines connected from floor supported IV equipment to the patient remain slack and there is no increase in the patient's heart rate.
These and other objects and advantages of the present invention will become more readily apparent during the following detailed description taken in conjunction with the drawings herein in which:
FIG. 1 is a perspective view of the patient weigh scale of the present invention docked to a hospital bed;
FIG. 2 is a perspective view of the bracing arm of the weigh scale;
FIGS. 3 and 4 illustrate the weigh scale being compactly configured for storage;
FIG. 5 is a perspective view of the latch mechanism of the weigh scale;
FIGS. 6 and 7 illustrate the rotational adjustability of the entire weigh scale and load cell, respectively;
FIG. 8 is a perspective view of the load cell mounting arrangement; and
FIG. 9 is a side view showing a schematic pivotal mounting between the crosspiece and boom.
With reference first to FIG. 1, there is illustrated a patient weigh scale 10 according to the present invention. The patient weigh scale 10 is shown removably secured to a hospital bed 20 having a base 22 which includes a longitudinal stem portion 24 a pair of outspread arms at one end of the stem portion, one of which is shown at 26, each arm 26 having an outwardly diverging portion 28 and a longitudinal portion 30, casters 32, frame 34 mounted for vertical movement relative to the base 22 via linkage 36, a patient support 38 supported by the frame 34 and raisable and lowerable sideguards 40. While scale 10 is shown removably secured to a particular side of bed 20, it will be appreciated that scale 10 could just as easily be removably secured to the other side of bed 20, and preferably bed 20 would be so equipped to allow for ambidextrous docking of scale 10 thereto. Patient support 38 is preferably substantially rigid so that when a patient supported by support 38 is suspended from scale 10, support 38 remains generally planar and does not "hammock" as traditional slings do.
Referring now to all the Figures, the patient weigh scale 10 includes a base 50 having casters 52, an upright support pole 54 mounted to the base 50, and a cantilever boom 56 pivotally mounted to an upper end of the upright support pole 54 via a bracket 58 and pin 60, which pin 60 provides a vertical swinging axis for boom 56.
A cross piece or member 62 is pivotally mounted to an outer end of the cantilever boom 60 for rotational movement about a generally vertical axis when the boom 56 is generally horizontal or perpendicular to the upright support pole 54 (FIG. 7 and FIG. 9) and as shown schematically by the rotation pin 140. At the ends of the cross member 62 there are mounted a pair of load cells 64, 64. Each load cell 64 has a fixed end 66 fixed to an end of the cross member 62 and a free end 68. The load cells 64 awe of the cantilever beam type. Suspended from the free ends 68 of the cantilever beam type load cells 64 is an H-frame 70 including arms 72, 72 and a connecting member 74. The free end 68 of the cantilever beams 64 are connected to the arms 72, 72 of the H-frame via links 76 and pins or other suitable fasteners 78, 80. Straps or other means 82 are provided for connecting the H-frame 70 to the lateral sides or edges of the patient support 38 and may in fact form a part of the patient support 38.
The bracing arm 84 includes sides 86 and 88 connected at one end by connector 90. At the lower end the bracing arm 84 is pivotally connected to the upright support post 54 via pivot 94. The upper ends of each arm 86 and 88 of the bracing arm 84 include through holes, one of which is shown at 96, which align with a similar through-hole (not shown) in the cantilever boom 56 when the boom 56 is generally perpendicular to the upright support pole 54. A pull pin 100 is removably securable through the holes for fixing the boom 56 in a general perpendicular relationship to the upright support pole 54 for weighing the patient, and may be removed from the bracing arm 84 and cantilever boom 56 thereby enabling the boom to fold downwardly into a generally parallel relationship with the pole 54 and into a compact storage position for storage purposes (FIGS. 3 and 4). Alternatively, the connector 90 can act as a pivot (similarly to pivot 94 in FIG. 2) with the pivot 94 becoming a pull pin aligned with holes in the sides 86 and 88 and pole 54 whereby the bottom end of the bracing arm can be disconnected and left to dangle from the boom 56 when the boom 56 is placed in the collapsed position. A knob 101 is mounted to the underside of the boom 56 against which connector 90 of arm 84 may be braced while inserting pin 100.
A combined handlebar and operator input/output display and entry device 102 is connected to the upright support pole 54. The handlebar aids in steering the scales, while the display and keypad entry device is for taking the weight of the patient, etc.
A latch 104 is mounted to the scales 10 and is for latching the scale 10 to the bed 20 so that during weighing of the patient supported by the bed there will be no relative movement between the scale 10 and bed 20. The latch 104 includes a split latch block 106 which encircles the upright support pole 54 and which is adjustably securable to the upright support pole 54 by conventional fasteners 107 so as to allow the pole 54 to rotate within the latch block 106. The block rests on the base 50. The fasteners 107 can be tightened to the point where the block 106 tightly grips and rotates with the pole 54. A pair of vertically planar plates 108, 108 are connected to the block 106 and project forwardly therefrom. The plates 108, 108 include a forwardly projecting lip 110 which resides beneath the lower edge of the portion 30 of the arm 28 when the scale 10 is docked to the bed 20. A pivoting latch finger 112 is pivoted to the upper portion of the plates 108, 108 via a pin 114 which defines a pivot axis. The latch finger 112 projects forwardly of the pivot axis and includes a ramped lower forward edge 116 forming a hook 118 therebehind. Ramp 116 is operable to ride upwardly and over a latch bar or pin 120 secured to a pair of ears 122, 122 which are mounted on the arm 26. A pair of rearwardly projecting foot pedals 124, 124 are connected to the latching finger 112. Depressing either of the foot pedals 124 causes the latching finger 112 to pivot upwardly about the pivot axis defined by pivot pin 114. Thus as the scale 10 is rolled against the bed base 22, lip 110 rides underneath of portion 30 of arm 26, latch is latching finger 112 rides upwardly and over pin 120 and snaps downwardly thereover to secure the scale 10 to the bed 20. The entire scale may then be rotated about a vertical axis within the block 106 (FIG. 6).
In use, the scale 10 is removed from its storage location, and the boom 56 is pivoted upwardly to a position generally perpendicular the upright support pole 54. Bracing arm 84 is then pivoted upwardly and connector 90 is placed behind knob 101; holes 96 are thus aligned with the holes in the boom 56. Pin 100 is then inserted therethrough fixing the boom in the horizontal operable position. The scale 10 is then rolled toward the hospital bed 20 via the handlebar 102 which supports the patient desired to be weighed. The base 50 is directed up towards the hospital bed base 22 so as to roll beneath the outspread arm 26. The latching finger 112 of the latch mechanism 104 is then aligned so as to pass between the lugs 122, 122 and to ride upwardly and over the latch pin or bar 120, while simultaneously the lip 110 of the plates 108, 108 travels beneath the arm 26.
Once the latch assembly 104 secures the scale 10 to the bed 20, the entire scale 20 may be rotated about a vertical axis (FIG. 6) so as to position the end of the boom 56 generally over the center of mass of a patient supported by the patient support 38. Once this is done, it may be necessary to rotate the cross member 62, load cells 64 and H-frame 70 relative to the boom 56 so as to square up that entire assembly relative to the patient support 38 (FIG. 7).
Once that is done, the straps or fastening means 82 are connected between the lateral sides or edges of the patient support 38 and the ends of the arms 72 of the H-frame 70. Next the frame 34 of the bed 20 is lowered via the frame mounting linkage 36 such that the patient support 38 is no longer supported by the frame 34, but is suspended entirely from the patient scale 10. At this point the patient's weight may be taken from the display 102.
Once the weight of the patient has been taken, the frame 34 is raised upwardly so as to support the patient atop the patient support 38. The straps 82 are disconnected, one of the pedals 124 is depressed so as to unlatch the scale 10 from the bed base 22, and the scale is then rolled via the handlebar 102 to its storage location, at which time pin 100 is withdrawn, arm 84 is pivoted downwardly and boom 56 with cross member 62 and H-frame 70 are pivoted downwardly into the compact storage position wherein the boom 56 is generally parallel the upright support pole 54.
The use of four load cells 64 substantially increases the accuracy of the scale 10 over prior art scales which traditionally employ only two load cells. Further, by employing a substantially rigid patient support 38 which remains generally planar even when suspending a patient from scale 10, orienting the load cells transversly to the patient support, suspending the H-frame 70 from the load cells and attaching the support to the ends of the arms 72 of the H-frame the "hammocking" tendency characteristic of prior art scales and slings is substantially reduced or eliminated, thus obviating the undesired raising of the patient's heart rate. Further, since the patient support remains at the same height relative to the floor during weighing, IV line pullout etc. does not occur as floor supported infusion equipment remains a fixed distance to the patient during weighing contrary to prior weighing methods which utilize raisable and lowerable boom scales.
Those skilled in the art will readily recognize numerous adaptations and modifications which can be made to the patient weigh scale of the present invention all of which will fall within the spirit and scope of the invention as defined in the following claims. Accordingly, the invention is to be limited only by the scope of the claims and their equivalents.
|Cited Patent||Filing date||Publication date||Applicant||Title|
|US1992262 *||11 May 1934||26 Feb 1935||Abe Dinkins||Folding portable packing and weighing stand|
|US2990899 *||24 Nov 1958||4 Jul 1961||Bella Isabelle D De||Bed patient weighing means|
|US3596725 *||1 Nov 1968||3 Aug 1971||Douglas Homs Corp||Hinged column for scale|
|US3795284 *||3 Jan 1972||5 Mar 1974||Bauer R||Portable support and weigher for a bed patient|
|US3876018 *||29 Nov 1973||8 Apr 1975||Said Mracek By Said Bauer||Portable support for a bed patient|
|US4006789 *||21 Jan 1976||8 Feb 1977||Acme Scale Company||Scale for weighing hospital patients in their horizontal position|
|US4033420 *||20 May 1976||5 Jul 1977||Brookline Instrument Company, Inc.||Weighing scale|
|US4281730 *||15 Jan 1980||4 Aug 1981||Swersey Burt L||Scale|
|US4420052 *||27 Aug 1980||13 Dec 1983||Scale-Tronix, Inc.||Patient weighing scale|
|US4482783 *||8 Mar 1976||13 Nov 1984||Hottinger Baldwin Measurements, Inc.||Patient weighing scale with hoist|
|US4487276 *||3 May 1983||11 Dec 1984||Swersey Burt L||Scale of flat construction|
|US4793428 *||29 Feb 1988||27 Dec 1988||Cobe Asdt, Inc.||Hospital bed with an integrated scale|
|US5033563 *||22 Jun 1981||23 Jul 1991||Cardinal Scale Manufacturing Company||Apparatus for weighing a patient on a stretcher|
|US5319817 *||15 Sep 1992||14 Jun 1994||Andermac, Inc.||Folding patient lift and weighing apparatus|
|Citing Patent||Filing date||Publication date||Applicant||Title|
|US5990423 *||12 Jun 1998||23 Nov 1999||Eze Weigh, Llc||Mobile hospital bed scale|
|US6374436||5 Sep 2000||23 Apr 2002||Hill-Rom Services, Inc.||Hospital bed|
|US6680443||19 Jun 2002||20 Jan 2004||Hill-Rom Services, Inc.||Load cell apparatus having a gap measuring device|
|US6694548||28 Feb 2002||24 Feb 2004||Hill-Rom Services, Inc.||Hospital bed|
|US6725474||16 Jul 2002||27 Apr 2004||Hill-Rom Services, Inc.||Hospital bed|
|US6924441||26 Sep 2000||2 Aug 2005||Hill-Rom Services, Inc.||Load cell apparatus|
|US6956175||6 Sep 2002||18 Oct 2005||Daly Paul C||Weighing apparatus and method|
|US7176391||24 Aug 2005||13 Feb 2007||Hill-Rom Services, Inc.||Load cell to frame interface for hospital bed|
|US7232962 *||20 Jan 2004||19 Jun 2007||Richard Rynd||Mobile hospital bed scale|
|US7335839||27 Dec 2006||26 Feb 2008||Hill-Rom Services, Inc.||Load cell interface for a bed having a stud receiver with a roller axis parallel with an axis of a load cell stud|
|US7459645||10 Dec 2004||2 Dec 2008||Hill-Rom Services, Inc.||Seat force sensor for a patient support|
|US7509698||8 Jan 2007||31 Mar 2009||Kreg Medical, Inc.||Therapeutic mattress|
|US7587776||10 Aug 2006||15 Sep 2009||Kreg Medical, Inc.||Dynamic therapy bed system|
|US7600434 *||30 Nov 2007||13 Oct 2009||Siemens Medical Solutions Usa, Inc.||Mechanism for remotely measuring medical treatment system bed deflection|
|US7644458||22 Jan 2007||12 Jan 2010||Hill-Rom Services, Inc.||Hospital bed|
|US7676862||12 Sep 2005||16 Mar 2010||Kreg Medical, Inc.||Siderail for hospital bed|
|US7714238||26 Nov 2008||11 May 2010||Hill-Rom Services, Inc.||Mattress seat force sensing method|
|US7716766||23 Mar 2009||18 May 2010||Kreg Medical, Inc.||Therapeutic mattress|
|US7743441||12 Sep 2005||29 Jun 2010||Kreg Therapeutics, Inc.||Expandable width bed|
|US7757318||20 Jul 2010||Kreg Therapeutics, Inc.||Mattress for a hospital bed|
|US7779494||12 Sep 2005||24 Aug 2010||Kreg Therapeutics, Inc.||Bed having fixed length foot deck|
|US8056160||5 Jan 2010||15 Nov 2011||Kreg Medical, Inc.||Siderail for hospital bed|
|US8069514||28 Jun 2010||6 Dec 2011||Kreg Medical, Inc.||Expandable width bed|
|US8090478||12 Jun 2006||3 Jan 2012||Hill-Rom Services, Inc.||Control for pressurized bladder in a patient support apparatus|
|US8112836||12 Feb 2009||14 Feb 2012||Linet Spol. S.R.O.||Positioning mechanism of a bed|
|US8397327||26 Aug 2011||19 Mar 2013||Span-America Medical Systems, Inc.||Bed insert|
|US8504323 *||15 Jul 2010||6 Aug 2013||Scott Anthony Coradi||Emergency patient medication system|
|US8620477||22 Dec 2011||31 Dec 2013||Hill-Rom Services, Inc.||Control for pressurized bladder in a patient support apparatus|
|US8717181||31 Jan 2011||6 May 2014||Hill-Rom Services, Inc.||Bed exit alert silence with automatic re-enable|
|US9107511||18 Dec 2013||18 Aug 2015||Hill-Rom Services, Inc.||Control for pressurized bladder in a patient support apparatus|
|US9119753||26 Jun 2009||1 Sep 2015||Kreg Medical, Inc.||Bed with modified foot deck|
|US20040020694 *||1 Aug 2002||5 Feb 2004||Hall Steven Ronald||In-bed human weighing device|
|US20040154842 *||20 Jan 2004||12 Aug 2004||Richard Rynd||Mobile hospital bed scale|
|US20060059814 *||24 Aug 2005||23 Mar 2006||Metz Darrell L||Load cell to frame interface for hospital bed|
|US20070070684 *||10 Aug 2006||29 Mar 2007||Craig Poulos||Dynamic therapy bed system|
|US20070107948 *||27 Dec 2006||17 May 2007||Metz Darrell L||Load cell to frame interface for hospital bed|
|US20070113352 *||8 Jan 2007||24 May 2007||Craig Poulos||Therapeutic mattress|
|US20070240912 *||2 Dec 2004||18 Oct 2007||Elias Papazissis||Bed-Top Scale|
|US20070272450 *||10 Dec 2004||29 Nov 2007||Hill-Rom Services, Inc.||Seat Force Sensor|
|US20090084609 *||26 Nov 2008||2 Apr 2009||Skinner Andrew F||Mattress seat force sensing method|
|US20090139032 *||30 Nov 2007||4 Jun 2009||Siemens Medical Solutions Usa, Inc.||Mechanism for Remotely Measuring Bed Deflection|
|US20110010858 *||12 Feb 2009||20 Jan 2011||Milan Tesar||Positioning mechanism of a bed|
|US20120016631 *||19 Jan 2012||Scott Anthony Coradi||Emergency patient medication system|
|U.S. Classification||177/144, 177/263|
|International Classification||A61G7/015, A61G7/10, A61G7/05, A61H3/04, G01G19/44, A61G5/14, A61G7/053, A61G7/02|
|Cooperative Classification||A61G7/012, G01G19/445, A61G7/108, A61H3/04, A61G7/1046, A61G7/1017, A61G7/1074, A61G7/1076, A61G7/05, A61G7/02, A61G7/1055|
|European Classification||G01G19/44A, A61H3/04, A61G7/10Z2, A61G7/10T6, A61G7/10Z8, A61G7/10N4|
|31 Mar 1994||AS||Assignment|
Owner name: HILL-ROM COMPANY, INC., INDIANA
Free format text: ASSIGNMENT OF ASSIGNORS INTEREST;ASSIGNORS:FOSTER, L. DALE;REEDER, RYAN A.;REEL/FRAME:006948/0684;SIGNING DATES FROM 19940316 TO 19940322
|29 Mar 2001||FPAY||Fee payment|
Year of fee payment: 4
|21 May 2001||AS||Assignment|
|20 Apr 2005||REMI||Maintenance fee reminder mailed|
|30 Sep 2005||LAPS||Lapse for failure to pay maintenance fees|
|29 Nov 2005||FP||Expired due to failure to pay maintenance fee|
Effective date: 20050930