The fine tuning of a sensor designed to locate interfaces using a two-dimensional gas–solid fluidized bed (FB) as reference is described. The sensor works through computer vision and consists of a charge coupled device camera that is placed alongside the bed and, as the human eye would do, first establishes a straight segment of the bed's top surface to define the phase boundary and then through pattern recognition continuously scans the interface in search of a similar pattern. The field of view is 582 × 752 pixels2. The experimentally measured Pixel size is 216 × 208 μm2. The device allows for experimentally obtaining the position of the probe with a resolution of ± 0.01 pixels at 25 Hz. We also describe its use to measure the height of the FB and observe a linear relationship with fluidization velocity. © 2012 American Institute of Chemical Engineers AIChE J, 2012
If you can't find a tool you're looking for, please click the link at the top of the page to "Go to old article view". Alternatively, view our Knowledge Base articles for additional help. Your feedback is important to us, so please let us know if you have comments or ideas for improvement.