The TX66A Series has been discontinued. Please see the TX91A and TX92A Series as a possible alternative or contact our Temperature Engineering Department.
The TX66 "smart" transmitter offers isolation, linearization, automatic temperature compensation, automatic self-calibration, and easy-to-use menus. These compact units accept a large number of inputs. The optional 1-line display will fit within the same footprint as the transmitter. The 2-line plug-in display is slightly larger, with two push buttons and LCD glass. Both displays are easily field mounted. At least one display is required to set up the transmitter.
SPECIFICATIONS Linearization: ±0.05°C Output: Analog: 2-wire Digital: 4 to 20 mA Minimum Output Range: None Output Rezolution: Analog: 3.5 µA Digital: 4 significant digits Accuracy: Digital: ±0.03% of the mV or equivalent reading or the accuracy from the table below, whichever is greater Analog: Digital accuracy plus ±4 µA, typical, ±8 µA max Repeatability: One half of accuracy Failsafe: User-programmable for high, low, or off Operating Temp Range: Transmitter: -40 to 85°C (-40 to 185°F) Display: -20 to 70°C (-4 to 158°F) 0 to 100% RH (non-condensing) Isolation: 500 Vdc or Vac p-p; operating input-output voltage: 250 Vac p-p or ±250 Vdc Power and Load: 12 to 48 Vdc; VSUPPLY (MIN) = 12 + (RLOAD in kΩ) x 23 mA Common-Mode Rejection: 120 dB Reverse Polarity Protection: 42 Vdc applied with either polarity Dimensions: 41 mm H x 61 Dia. (1.6 x 2.4") Weight: 227 g (8 oz)
† All amounts shown in USD Note: Comes with complete operator’s manual