Publication: Research - peer-review › Conference abstract for conference – Annual report year: 2010
Current ultrasound scanners can only estimate the velocity along the ultrasound beam and this gives rise to the cos() factor on all velocity estimates. This is a major limitation as most vessels are close to perpendicular to the beam. Also the angle varies as a function of space and time making it virtually impossible to compensate for the factor and obtain correct velocity estimates for either CFM or spectral velocity estimation. This talk will describe methods for finding the correct velocity by estimating both the axial and lateral component of the velocity vector. The transverse oscillation method introduces an ultrasound field that oscillation not only along the ultrasound beam both also transverse to it to estimate both the lateral and axial velocity for the full velocity vector. The correct velocity magnitude can be found from this as well as the instantaneous angle. This can be obtained over the full region of interest and a real time image at a frame rate of 20 Hz can be displayed. Real time videos have been obtained from both our research systems and from commercial BK Medical scanners. The vector velocity images reveal the full complexity of the human blood flow. It is easy to see direction and the correct velocity magnitude for any orientation of the vessels. At complex geometries like bifurcations, branching and for valves the approach reveals how the velocity changes magnitude and direction over the cardiac cycle. Vector velocity reveals a wealth of new information that now is accessible to the ultrasound community. The displaying and studying of this information is challenging as complex flow changes rapidly over the cardiac cycle.
|State||Published - 2010|
|Conference||8th European Conference on Synthetic Aperture Radar|
|Period||07/06/2010 → 10/06/2010|
Loading map data...
No data available