DTSD1088H3 CDMA meter,
Three phase electronic multifunction watt hour meter
The key features of the DT(S)SD1088H3-series meter (3P4W) are cost-effective multi-functional CDMA metering device for commercial and industrial applications. The meter offers the powerful functions such as TOU (Time-of-Use) metering, various energy and power measurements, huge metering/interval data and events recording, instantaneous measurement and instrumentation, tamper & fraud protection, and communication.
2. Metering capabilities
The meter can measure and record a billing data like energy and demand data based on a metering channel configuration which can set up to 4-channels. Also, the meter can measure the quadrant energy and the combination of each quadrant energy quantities. For more requirements, the demand measuring and recording channel can be set separately with the energy metering and recording channel.
1). TOU metering
The meter can adopt up to 8-tariff TOU metering capabilities including 8-switching points per day, 12-day schedules, 8-week groups, and more than 600-holidays. Additionally, the meter provides a unique TOU function that each metering and recording channels can be set to be an exclusive TOU program.
2). Load profile
3). Instrumentation profile (load survey)
5). Power supply
6). Event log
8). Tamper& Fraud Protection
The brief specification of the DT(S)SD1088H3 series meter is following:
Voltage: AC 220/380V;
Current: 5(10) A;
Frequency Nominal: more or less 5 % than 50Hz;
Temperature: -25 ~ 75 degree;
Humidity: Up to 100% non-condensing Power loss Less than 2W;
Accuracy: 0.5 / 1.0 class;
Accuracy variations: Voltage coefficient is more or less than0.03%;
Temperature coefficient: More or less than 0.03% per degree;
Starting current: Less than 0.005 In;
Startup delay: Less than 3 sec;
Creep: No more than 1-pulse;
Communication: 300~9600 baud;
Standards: IEC 62052-11, IEC 62053-21, 22, 23, IEC 62056-21.