The new Zetasizer APS (Auto Plate Sampler) from Malvern
Instruments delivers fully automated measurement of protein size, with no
compromise in data accuracy or quality. An ideal solution to improve productivity,
this dynamic light scattering (DLS) system handles 96- or 384-well microplates,
transferring the sample to a closely controlled environment for robust size
characterization. Measurements within this defined environment provide the accuracy
and temperature stability needed to deliver consistent, reliable results. The
resulting data precision gives access to more applications, particularly those
requiring the detection of small amounts of aggregates or the early onset of
In dynamic light scattering measurements, temperature is a critical parameter.
A one degree uncertainly in the temperature can mean a 5% range of uncertainty
in the result. The Zetasizer APS automatically extracts a sample directly from
its well and then measures it within an enclosed measurement area that operates
in the range 2 to 90°C, controlled to 0.1 degree Celsius. Dual temperature
controls maintain optimal conditions within the wellplate to preserve sample
integrity, and separately within the measurement chamber. Each sample on a plate
can be measured using a different protocol and each is preserved for further
investigation. Melting point measurements can be carried out on individual samples,
each using a different range of temperatures, without affecting the other samples
in the plate.
Since the Zetasizer APS can handle virtually all commercially available microwell
plates, no special materials are needed. This avoids the use of expensive plates
with special optical properties, where re-use can be tempting but then introduces
the risk of cross-contamination. A programmable system of rinsing the extraction
needle and tubes before and after measurement also ensures no sample cross-contamination.
Two fluid inputs provide two separately controlled rinses: a solvent and a wash.
A background measurement between every sample means that results are thoroughly
tested for extra confidence.
Automation of size measurements using the Zetasizer APS enables unattended
measurement of a large number of samples without sacrificing data quality. The
temperature controlled measurement chamber allows measurement in the range of
2 to 90°C with a 0.1°C precision. Easy-to-use software provides a range
of result reports, as well as the ability to customise data output.