|Publication number||US7651492 B2|
|Application number||US 11/409,574|
|Publication date||26 Jan 2010|
|Filing date||24 Apr 2006|
|Priority date||24 Apr 2006|
|Also published as||CA2585622A1, DE602007002940D1, EP1849425A1, EP1849425B1, EP2140825A1, EP2140825B1, EP2298208A1, US8556890, US20070250052, US20100094275, US20140039490|
|Publication number||11409574, 409574, US 7651492 B2, US 7651492B2, US-B2-7651492, US7651492 B2, US7651492B2|
|Inventors||Robert H. Wham|
|Original Assignee||Covidien Ag|
|Export Citation||BiBTeX, EndNote, RefMan|
|Patent Citations (101), Non-Patent Citations (61), Referenced by (9), Classifications (9), Legal Events (3) |
|External Links: USPTO, USPTO Assignment, Espacenet|
Arc based adaptive control system for an electrosurgical unit
US 7651492 B2
A system and method for performing electrosurgical procedures are disclosed. The system includes an electrosurgical generator adapted to supply electrosurgical energy to tissue in form of one or more electrosurgical waveforms having a crest factor and a duty cycle. The system also includes sensor circuitry adapted to measure impedance and to obtain one or more measured impedance signals. The sensor circuitry is further adapted to generate one or more arc detection signals upon detecting an arcing condition§. The system further includes a controller adapted to generate one or more target control signals as a function of the measured impedance signals and to adjust output of the electrosurgical generator based on the arc detection signal. An electrosurgical instrument is also included having one or more active electrodes adapted to apply electrosurgical energy to tissue.
1. An electrosurgical system comprising:
an electrosurgical generator adapted to supply electrosurgical energy to tissue in the form of at least one electrosurgical waveform having a crest factor and a duty cycle;
sensor circuitry adapted to measure impedance and to obtain at least one measured impedance signal, the sensor circuitry further adapted to generate at least one arc detection signal upon detection of an arcing condition;
a high pass filter adapted to filter the at least one measured impedance signal to obtain an absolute value thereof;
a low pass filter adapted to filter the absolute value of the at least one measured impedance signal to obtain the at least one arc detection signal;
a controller adapted to generate at least one target control signal as a function of the at least one measured impedance signal and to adjust output of the electrosurgical generator based on the at least one arc detection signal; and
an electrosurgical instrument including at least one active electrode adapted to apply electrosurgical energy to tissue.
2. An electrosurgical system as in claim 1, wherein the at least one arc detection signal is scaled to represent the level of arcing.
3. An electrosurgical system as in claim 1, wherein the controller adjusts the output of the electrosurgical generator by generating at least one target control signal as a function of an average impedance value.
4. An electrosurgical system as in claim 1, wherein the controller adjusts output of the electrosurgical generator by adjusting at least one of the crest factor and the duty cycle of the at least one waveform.
5. A method for performing an electrosurgical procedure comprising the steps of:
supplying electrosurgical energy from an electrosurgical generator to tissue in the form of at least one electrosurgical waveform having a crest factor and a duty cycle;
measuring impedance to obtain at least one measured impedance signal;
generating at least one arc detection signal upon detecting an arcing conditions;
filtering the at least one measured impedance signal through a high pass filter to obtain an absolute value thereof;
filtering the absolute value of the at least one measured impedance signal through a low pass filter to obtain the at least one arc detection signal;
generating at least one target control signal as a function of the at least one measured impedance signal; and
adjusting output of the electrosurgical generator based on the at least one arc detection signal.
6. A method as in claim 5
, further comprising the step of:
scaling the at least one arc detection signal according to a scale representing level of arcing.
7. A method as in claim 5
, wherein the step of adjusting output of the electrosurgical generator further comprises:
adjusting the output of the electrosurgical generator by generating at least one target control signal as a function of an average impedance value.
8. A method as in claim 5
, wherein the step of adjusting output of the electrosurgical generator further comprises:
adjusting at least one of the crest factor and the duty cycle of the at least one waveform.
9. An electrosurgical generator:
an RF output stage adapted to supply electrosurgical energy to tissue in the form of at least one electrosurgical waveform having a crest factor and a duty cycle;
sensor circuitry adapted to measure impedance and to obtain at least one measured impedance signal, the sensor circuitry further adapted to generate at least one arc detection signal upon detecting an arcing condition;
a high pass filter adapted to filter the at least one measured impedance signal to obtain an absolute value thereof;
a low pass filter adapted to filter the absolute value of the at least one measured impedance signal to obtain the at least one arc detection signal; and
a controller adapted to generate at least one target control signal as a function of the at least one measured impedance signal and to adjust output of the electrosurgical generator based on the at least one arc detection signal.
10. An electrosurgical generator as in claim 9, wherein the at least one arc detection signal is scaled according to represent the level of arcing.
11. An electrosurgical generator as in claim 9, wherein the controller adjusts the output of the electrosurgical generator by generating at least one target control signal as a function of an average impedance value.
12. An electrosurgical generator as in claim 9, wherein the controller adjusts output of the electrosurgical generator by adjusting at least one of the crest factor and the duty cycle of the at least one waveform.
1. Technical Field
The present disclosure relates to electrosurgical generators. More particularly, the present disclosure relates to a system and method for controlling output of an electrosurgical generator. The electrosurgical generator includes a sensing feedback control system and an arc-based adaptive control system which adjusts output in response to arcing.
2. Background of Related Art
Energy based tissue treatment is well known in the art. Various types of energy (e.g., electrical, ultrasonic, microwave, cryo, heat, laser, etc.) are applied to tissue to achieve a desired result. Electrosurgery involves application of high radio frequency electrical current to a surgical site to cut, ablate, coagulate send or otherwise seal tissue. In monopolar electrosurgery, a source or active electrode delivers radio frequency energy from the electrosurgical generator to the tissue and a return electrode carries the current back to the generator. In monopolar electrosurgery, the source electrode is typically part of the surgical instrument held by the surgeon and applied to the tissue to be treated. A patient return electrode is placed remotely from the active electrode to carry the current back to the generator.
In bipolar electrosurgery, one of the electrodes of the hand-held instrument functions as the active electrode and the other as the return electrode. The return electrode is placed in close proximity to the active electrode such that an electrical circuit is formed between the two electrodes (e.g., electrosurgical forceps). In this manner, the applied electrical current is limited to the body tissue positioned between the electrodes. When the electrodes are sufficiently separated from one another, the electrical circuit is open and thus inadvertent contact with body tissue with either of the separated electrodes does not cause current to flow.
It is known in the art that sensed tissue feedback may be used to control delivery of electrosurgical energy. During application of electrosurgical energy, arcing may occur during the course of treatment. Energy arcing is particularly problematic for sensed feedback control systems since the systems attempt to adjust to the rapidly occurring changes in tissue properties caused by arcing.
According to one aspect of the present disclosure an electrosurgical system is disclosed. The system includes an electrosurgical generator adapted to supply electrosurgical energy to tissue in form of one or more electrosurgical waveforms having a crest factor and a duty cycle. The system also includes sensor circuitry adapted to measure impedance and to obtain one or more measured impedance signals. The sensor circuitry is further adapted to generate one or more arc detection signals upon detecting an arcing condition. The system further includes a controller adapted to generate one or more target control signals as a function of the measured impedance signals and to adjust output of the electrosurgical generator based on the arc detection signal. An electrosurgical instrument is also included having one or more active electrodes adapted to apply electrosurgical energy to tissue.
Another aspect of the present disclosure includes a method for performing an electrosurgical procedure. The method includes the steps of: supplying electrosurgical energy from an electrosurgical generator to tissue in form of one or more electrosurgical waveforms having a crest factor and a duty cycle and measuring impedance to obtain one or more measured impedance signals and generating one or more detection signal upon detecting arcing conditions. The method further includes the steps of generating one or more target control signals as a function of the measured impedance signals and adjusting output of the electrosurgical generator based on the arc detection signals.
According to a further aspect of the present disclosure an electrosurgical generator is disclosed. The generator includes an RF output stage adapted to supply electrosurgical energy to tissue in form of one or more electrosurgical waveforms having a crest factor and a duty cycle. The generator also includes sensor circuitry adapted to measure impedance and to obtain one or more measured impedance signals. The sensor circuitry is further adapted to generate one or more arc detection signals upon detecting one or more arcing conditions. The generator further includes a controller adapted to generate one or more target control signals as a function of the measured impedance signals and to adjust output of the electrosurgical generator based on the arc detection signal
BRIEF DESCRIPTION OF THE DRAWINGS
Various embodiments of the present disclosure are described herein with reference to the drawings wherein:
FIG. 1 is a schematic block diagram of an electrosurgical system according to the present disclosure;
FIG. 2 is a schematic block diagram of a generator according to the present disclosure; and
FIG. 3 is a flow diagram illustrating a method according to the present disclosure.
Particular embodiments of the present disclosure are described hereinbelow with reference to the accompanying drawings. In the following description, well-known functions or constructions are not described in detail to avoid obscuring the present disclosure in unnecessary detail. Those skilled in the art will understand that the method according to the present disclosure may be adapted to monitor use with either monopolar or bipolar electrosurgical systems.
FIG. 1 is a schematic illustration of an electrosurgical system according to the present disclosure. The system includes an electrosurgical instrument 10 having one or more electrodes for treating tissue of a patient P. The instrument 10 may be either a monopolar type including one or more active electrodes (e.g., electrosurgical cutting probe, ablation electrode(s), etc.) or a bipolar type including one or more active and return electrodes (e.g., electrosurgical sealing forceps). Electrosurgical RF energy is supplied to the instrument 10 by a generator 20 via a supply line 12, which is operably connected to an active output terminal, allowing the instrument 10 to coagulate, seal, ablate and/or otherwise treat tissue.
If the instrument 10 is a monopolar type instrument then energy may be returned to the generator 20 through a return electrode (not explicitly shown) which may be disposed on the patient's body. The system may also include a plurality of return electrodes which are arranged to minimize the chances of damaged tissue by maximizing the overall contact area with the patient P. In addition, the generator 20 and the monopolar return electrode may be configured for monitoring so called “tissue-to-patient” contact to insure that sufficient contact exists therebetween to further minimize chances of tissue damage.
If the instrument 10 is a bipolar type instrument, the return electrode is disposed in proximity to the active electrode (e.g., on opposing jaws of a bipolar forceps). It is also envisioned that the generator 20 may include a plurality of supply and return terminals and a corresponding number of electrode leads.
The generator 20 includes input controls (e.g., buttons, activators, switches, touch screen, etc.) for controlling the generator 20. In addition, the generator 20 may include one or more display screens for providing the surgeon with a variety of output information (e.g., intensity settings, treatment complete indicators, etc.). The controls allow the surgeon to adjust power of the RF energy, waveform, and other parameters to achieve the desired waveform suitable for a particular task (e.g., coagulating, tissue sealing, intensity setting, etc.). It is also envisioned that the instrument 10 may include a plurality of input controls which may be redundant with certain input controls of the generator 20. Placing the input controls at the instrument 10 allows for easier and faster modification of RF energy parameters during the surgical procedure without requiring interaction with the generator 20.
FIG. 2 shows a schematic block diagram of the generator 20 having a controller 24, a high voltage DC power supply 27 (“HVPS”) and an RF output phase 28. The HVPS 27 provides high voltage DC power to an RF output phase 28 which then converts high voltage DC power into RF energy and delivers the high frequency RF energy to the active electrode 24. In particular, the RF output phase 28 generates sinusoidal waveforms of high frequency RF energy. The RF output phase 28 is configured to generate a plurality of waveforms having various duty cycles, peak voltages, crest factors, and other parameters. Certain types of waveforms are suitable for specific electrosurgical modes. For instance, the RF output phase 28 generates a 100% duty cycle sinusoidal waveform in cut mode, which is best suited for dissecting tissue and a 25% duty cycle waveform in coagulation mode, which is best used for cauterizing tissue to stop bleeding.
The controller 24 includes a microprocessor 25 operably connected to a memory 26 which may be volatile type memory (e.g., RAM) and/or non-volatile type memory (e.g., flash media, disk media, etc.). The microprocessor 25 includes an output port which is operably connected to the HVPS 27 and/or RF output phase 28 allowing the microprocessor 25 to control the output of the generator 20 according to either open and/or closed control loop schemes.
A closed loop control scheme or feedback control loop is provided that includes sensor circuitry 22 having one or more sensors for measuring a variety of tissue and energy properties (e.g., tissue impedance, tissue temperature, output current and/or voltage, etc.). The sensor circuitry 22 provides feedback to the controller 24. Such sensors are within the purview of those skilled in the art. The controller 24 then signals the HVPS 27 and/or RF output phase 28 which then adjust DC and/or RF power supply, respectively. The controller 24 also receives input signals from the input controls of the generator 20 or the instrument 10. The controller 24 utilizes the input signals to adjust power outputted by the generator 20 and/or performs other control functions thereon.
In particular, sensor circuitry 22 is adapted to measure tissue impedance. This is accomplished by measuring voltage and current signals and calculating corresponding impedance values as a function thereof at the sensor circuitry 22 and/or at the microprocessor 25. Power and other energy properties may also be calculated based on collected voltage and current signals. The sensed impedance measurements are used as feedback by the generator 20 for regulating the energy delivery to the tissue. Various types of impedance feedback control schemes are envisioned, such as for example, impedance matching (wherein power output is adjusted to match measured impedance to target impedance), impedance maintenance (epower is adjusted to maintain impedance), etc.
The sensor circuitry 22 obtains impedance signals and sends a corresponding target control signal that is used by the controller 24 to adjust the output of the generator 20. Since the output of the generator 20 is adjusted as a function of the measured impedance, the output of the generator 20 responds to every fluctuation in the measured impedance. While high resolution response times are desirable, certain variations in impedance may not warrant adjustments to the output of the generator 20.
During electrosurgical procedures, it is known that arcing causes rapid changes in impedance. Impedance fluctuates between arc impedance to open circuit impedance (e.g., nil). An impedance feedback control scheme that adjusts the output of the generator 20 in response to impedance changes for each arc is undesirable. If the output of the generator 20 tracks the impedance changes caused by arcing too closely, unwanted oscillation in the output may occur. The present disclosure provides an arc-based adaptive control method for mitigating oscillation and adjusting the behavior of an impedance feedback control scheme of the controller 24.
FIG. 3 shows an arc-based adaptive control method according to one embodiment of the present disclosure which is configured to control the output of the generator in response to monitored tissue impedance. In step 100, the instrument 10 engages the tissue and the generator 20 supplies electrosurgical energy to the tissue through the instrument 10. In step 110, during application of energy to the tissue, impedance is continually monitored by the sensor circuitry 22 and a measured impedance signal is obtained. As discussed above, the measured impedance signal is derived from voltage and current signals.
In step 120, a target control signal is generated by the controller 24 as a function of the measured impedance signal. In particular, the target control signal is generated by using output control algorithms which may operate in a wide variety of ways. For example, the algorithms may attempt to match measured impedance signal to predetermined target impedance or may simply use look-up tables containing corresponding target control signals. The output control algorithms are stored within the memory 26 and are executed by the microprocessor 26. Consequently, the target control signal is used to make appropriate adjustments to the output of the generator 20.
In step 130, arcing is detectable by monitoring for rapid repeating changes in measured impedance signal, target control signal, or voltage and current signals. Since target control signal and voltage and current signals are directly related to measured impedance signal, rapid changes in those signals are also indicative of arcing conditions. In other words, impedance correlates with arcing—low impedance is measured during an arc condition followed by high impedance when arcing stops. In step 140, an arc detection signal is generated. With reference to FIG. 2, this is accomplished by passing the measured impedance tissue signal through a high pass filter 29 and then pass the absolute value of the high pass through a low pass filter 30. The resulting filtered signal is the arc detection signal that is scaled and capped (for example by a 0 to 1 scale representing the level of arcing, where 1 represents heavy arcing and 0 represents no arcing). The arc detection signal rises as arcing increases and reduces and arcing decreases.
Arcing may also be detected by detecting rapid changes in the target control signal and/or by detecting rapid changes in either the voltage or current signals. If arcing is detected, the method proceeds to step 150 wherein the arc detection signal is used to make adjustments to the output control algorithm of the controller 24. This may be achieved by substituting the measured impedance signal with an average impedance value. By using the average impedance value to obtain the target control signal the generator 20 avoids using extreme impedance values associated with arcing. Alternatively, the output control algorithm may be configured to include selecting the target control signal associated with either the minimum or maximum impedance measured during arcing. This selects an impedance signal that is closest to the previously measured impedance ensuring that the impedance signal and, hence, the target control signal, do not deviate substantially from other values. Upon detecting arcing, the controller 24 may also stop and/or hold from issuing any target control signals thereby maintaining the output based on an immediately preceding measured impedance signal.
In response to arcing, RF generation may be stopped to quickly remove the arcing condition. This may be achieved to shutting down the RF output stage 28 and/or the HVPS 27. In addition, the circuit between the patient and generator 20 may be opened to prevent the RF energy from reaching tissue.
Other optimizations to the controller 24 are envisioned so that arcing is extinguished or energy delivery is enhanced during arcing. In particular, the calculations performed by the output control algorithm may be modified so that the desired output voltage, current and/or power are adjusted when arcing is present.
Further, modification to the waveform produced by the RF output stage 28 may be made in response to arcing. This may include momentary modification of the crest factor (i.e., ratio of the peak value to RMS value), the waveform and/or the duty cycle of the waveform. Adjustments to the crest factor and the duty cycle enhance or extinguish the arc. Low duty cycles tend to provide coagulation behavior whereas high duty cycles tend to provide for better cutting behavior. Thus, momentarily increasing the duty cycle of the waveform extinguishes the arc. Other ways of providing arc-based adaptive control include adjusting PD gain of the controller 24 such that the gain is reduced as the level of arcing increases.
In step 160, after the arcing conditions are removed, the modifications made to the controller 24 and/or the waveform are recalibrated and normal operating conditions are restored.
The arc-based adaptive control method according to the present disclosure allows for fast response to changes in non-arcing conditions and slows down the controller 24 during arcing so that the controller may readily select an alternate output value for heating a particular tissue type. This limits aggressive arcing behavior (e.g., so called entry/exit sparking). Another recognizable advantage is that electrosurgical systems utilizing arc-based control methods are capable of automatically switching from cutting to coagulation mode. Thus, during low arcing conditions, the system is optimized for cutting, but as the arcing increases the system adjusts so that coagulation is enhanced.
While several embodiments of the disclosure have been shown in the drawings and/or discussed herein, it is not intended that the disclosure be limited thereto, as it is intended that the disclosure be as broad in scope as the art will allow and that the specification be read likewise. Therefore, the above description should not be construed as limiting, but merely as exemplifications of particular embodiments. Those skilled in the art will envision other modifications within the scope and spirit of the claims appended hereto.
|Cited Patent||Filing date||Publication date||Applicant||Title|
|US1787709||11 Jun 1928||6 Jan 1931||Charles Wappler Frederick||High-frequency surgical cutting device|
|US1813902||18 Jan 1928||14 Jul 1931||Liebel Flarsheim Co||Electrosurgical apparatus|
|US1841968||16 Aug 1924||19 Jan 1932||William J Cameron||Radio-surgical apparatus|
|US1863118||31 Oct 1927||14 Jun 1932||Liebel Flarsheim Co||Surgeon's instrument|
|US1945867||27 Apr 1932||6 Feb 1934||Technical Equipment Company||High frequency oscillatory apparatus for electrotherapeutic and sterilization purposes|
|US2827056||21 Jun 1955||18 Mar 1958||John H Drew||Electrode discharge control for surgical apparatus|
|US2849611||16 May 1955||26 Aug 1958||Honeywell Regulator Co||Electrical oscillator circuit|
|US3058470||19 Apr 1957||16 Oct 1962||Siemens Reiniger Werke Ag||Apparatus for electrical highfrequency surgery|
|US3089496||19 Aug 1959||14 May 1963||Code Inc||Control system for surgical apparatus|
|US3154365||16 Jul 1962||27 Oct 1964||Kent Mfg Co||Holder for conductor tape|
|US3163165||12 Sep 1960||29 Dec 1964||Humio Islkawa||Uterotube-closing instrument|
|US3252052||23 Aug 1963||17 May 1966||Jacuzzi Bros Inc||Leakage detection and control circuit|
|US3391351||21 Nov 1966||2 Jul 1968||Bell Telephone Labor Inc||Circuits using a transistor operated into second breakdown region|
|US3413480||29 Nov 1963||26 Nov 1968||Texas Instruments Inc||Electro-optical transistor switching device|
|US3436563||27 Dec 1965||1 Apr 1969||Bell Telephone Labor Inc||Pulse driver with linear current rise|
|US3439253||5 Apr 1967||15 Apr 1969||R I Phelon Inc||Alternator rectifier and voltage regulator|
|US3439680||12 Apr 1965||22 Apr 1969||Univ Northwestern||Surgical instrument for cataract removal|
|US3461874||10 Aug 1966||19 Aug 1969||Martinez Miguel||Electric cautery|
|US3471770||30 Mar 1966||7 Oct 1969||Ibm||Pulsed current generating circuits|
|US3478744||30 Dec 1964||18 Nov 1969||Leiter Harry||Surgical apparatus|
|US3486115||31 Mar 1966||23 Dec 1969||Anderson Donald J||Means for measuring the power in an electrical circuit|
|US3495584||3 Jun 1965||17 Feb 1970||Gen Electric||Lead failure detection circuit for a cardiac monitor|
|US3513353||17 Aug 1967||19 May 1970||Lansch John L||Voltage monitoring circuit|
|US3514689||21 Aug 1968||26 May 1970||United Aircraft Corp||Three-phase ac-operated dc power supply|
|US3515943||29 Sep 1967||2 Jun 1970||English Electric Co Ltd||Electrical fault detector|
|US3551786||5 Dec 1967||29 Dec 1970||Omark Industries Inc||Circuit for adjustably increasing or decreasing the charge on a capacitor|
|US3562623||16 Jul 1968||9 Feb 1971||Hughes Aircraft Co||Circuit for reducing stray capacity effects in transformer windings|
|US3571644||27 Jan 1969||23 Mar 1971||Heurtey Sa||High frequency oscillator for inductive heating|
|US3589363||25 Jul 1967||29 Jun 1971||Cavitron Corp||Material removal apparatus and method employing high frequency vibrations|
|US3595221||4 Mar 1969||27 Jul 1971||Matburn Holdings Ltd||Endoscopic having illumination supply unit|
|US3601126||8 Jan 1969||24 Aug 1971||Electro Medical Systems Inc||High frequency electrosurgical apparatus|
|US3611053||10 Oct 1969||5 Oct 1971||Farmer Electric Products Co In||Intrinsically safe circuit|
|US3641422||1 Oct 1970||8 Feb 1972||Mccaughey Dennis G||Wide band boost regulator power supply|
|US3642008||15 Oct 1969||15 Feb 1972||Medical Plastics Inc||Ground electrode and test circuit|
|US3662151||17 Nov 1969||9 May 1972||Codman & Shurtleff||Cautery|
|US3675655||4 Feb 1970||11 Jul 1972||Electro Medical Systems Inc||Method and apparatus for high frequency electric surgery|
|US3683923||25 Sep 1970||15 Aug 1972||Valleylab Inc||Electrosurgery safety circuit|
|US3693613||9 Dec 1970||26 Sep 1972||Cavitron Corp||Surgical handpiece and flow control system for use therewith|
|US3697808||23 Nov 1970||10 Oct 1972||Safety Co The||System for monitoring chassis potential and ground continuity|
|US3699967||30 Apr 1971||24 Oct 1972||Valleylab Inc||Electrosurgical generator|
|US3720896||18 May 1971||13 Mar 1973||Siemens Ag||Handle for high frequency electrodes|
|US3743918||31 Jan 1972||3 Jul 1973||Philips Corp||Dc-ac converter|
|US3766434||9 Aug 1971||16 Oct 1973||Sherman S||Safety power distribution system|
|US3768482||10 Oct 1972||30 Oct 1973||R Shaw||Surgical cutting instrument having electrically heated cutting edge|
|US3801766||22 Jan 1973||2 Apr 1974||Valleylab Inc||Switching means for an electro-surgical device including particular contact means and particular printed-circuit mounting means|
|US3801800||26 Dec 1972||2 Apr 1974||Valleylab Inc||Isolating switching circuit for an electrosurgical generator|
|US3812858||24 Oct 1972||28 May 1974||Sybron Corp||Dental electrosurgical unit|
|US3815015||20 Feb 1973||4 Jun 1974||Gen Electric||Transformer-diode isolated circuits for high voltage power supplies|
|US3826263||7 Aug 1972||30 Jul 1974||R Shaw||Electrically heated surgical cutting instrument|
|US3848600||30 Jan 1973||21 Jun 1988|| ||Title not available|
|US3870047||12 Nov 1973||11 Mar 1975||Dentsply Res & Dev||Electrosurgical device|
|US3875945||2 Nov 1973||8 Apr 1975||Demetron Corp||Electrosurgery instrument|
|US3885569||21 Nov 1972||27 May 1975||Birtcher Corp||Electrosurgical unit|
|US3897787||27 Dec 1973||5 Aug 1975||Olympus Optical Co||Power source device for an electric surgical knife|
|US3897788||14 Jan 1974||5 Aug 1975||Valleylab Inc||Transformer coupled power transmitting and isolated switching circuit|
|US3898554||14 Nov 1973||5 Aug 1975||Danfoss As||Measured-value transducer with a compensating bridge circuit|
|US3905373||18 Apr 1974||16 Sep 1975||Dentsply Res & Dev||Electrosurgical device|
|US3913583||3 Jun 1974||21 Oct 1975||Sybron Corp||Control circuit for electrosurgical units|
|US3923063||15 Jul 1974||2 Dec 1975||Sybron Corp||Pulse control circuit for electrosurgical units|
|US3933157||23 Oct 1973||20 Jan 1976||Aktiebolaget Stille-Werner||Test and control device for electrosurgical apparatus|
|US3946738||24 Oct 1974||30 Mar 1976||Newton David W||Leakage current cancelling circuit for use with electrosurgical instrument|
|US3952748||18 Jul 1974||27 Apr 1976||Minnesota Mining And Manufacturing Company||Electrosurgical system providing a fulguration current|
|US3963030||1 Nov 1974||15 Jun 1976||Valleylab, Inc.||Signal generating device and method for producing coagulation electrosurgical current|
|US3964487||9 Dec 1974||22 Jun 1976||The Birtcher Corporation||Uncomplicated load-adapting electrosurgical cutting generator|
|US3971365||12 Feb 1973||27 Jul 1976||Beckman Instruments, Inc.||Bioelectrical impedance measuring system|
|US3978393||21 Apr 1975||31 Aug 1976||Burroughs Corporation||High efficiency switching regulator|
|US3980085||10 Jun 1975||14 Sep 1976||Olympus Optical Co., Ltd.||High frequency apparatus for heat treatment of biological tissue|
|US4005714||30 Jul 1975||1 Feb 1977||Richard Wolf Gmbh||Bipolar coagulation forceps|
|US4024467||2 Jun 1975||17 May 1977||Sybron Corporation||Method for controlling power during electrosurgery|
|US4041952||4 Mar 1976||16 Aug 1977||Valleylab, Inc.||Electrosurgical forceps|
|US4051855||6 Feb 1976||4 Oct 1977||Ipco Hospital Supply Corporation, Whaledent International Division||Electrosurgical unit|
|US4074719||24 Jun 1976||21 Feb 1978||Kurt Semm||Method of and device for causing blood coagulation|
|US4092986||14 Jun 1976||6 Jun 1978||Ipco Hospital Supply Corporation (Whaledent International Division)||Constant output electrosurgical unit|
|US4094320||9 Sep 1976||13 Jun 1978||Valleylab, Inc.||Electrosurgical safety circuit and method of using same|
|US4097773||1 Nov 1976||27 Jun 1978||Lindmark Magnus C W||Switched mode power supply|
|US4102341||16 Dec 1976||25 Jul 1978||Olympus Optical Co., Ltd.||Electric knife device|
|US4114623||29 Jul 1976||19 Sep 1978||Karl Storz Endoscopy-America, Inc.||Cutting and coagulation apparatus for surgery|
|US4121590||14 Mar 1977||24 Oct 1978||Dentsply Research And Development Corporation||System for monitoring integrity of a patient return circuit|
|US4123673||14 Mar 1977||31 Oct 1978||Dentsply Research And Development Corporation||Control circuit for an electrical device|
|US4126137||21 Jan 1977||21 Nov 1978||Minnesota Mining And Manufacturing Company||Electrosurgical unit|
|US4171700||13 Oct 1977||23 Oct 1979||Erbe Elektromedizin Gmbh & Co. Kg||High-frequency surgical apparatus|
|US4188927||12 Jan 1978||19 Feb 1980||Valleylab, Inc.||Multiple source electrosurgical generator|
|US4191188||9 Sep 1978||4 Mar 1980||Macan Engineering & Manufacturing Company, Inc.||Variable crest factor high frequency generator apparatus|
|US4196734||16 Feb 1978||8 Apr 1980||Valleylab, Inc.||Combined electrosurgery/cautery system and method|
|US4200104||17 Nov 1977||29 Apr 1980||Valleylab, Inc.||Contact area measurement apparatus for use in electrosurgery|
|US4200105||26 May 1978||29 Apr 1980||Dentsply Research & Development Corp.||Electrosurgical safety circuit|
|US4209018||30 May 1978||24 Jun 1980||Karl Fastenmeier||Tissue coagulation apparatus and method|
|US4231372||4 Nov 1974||4 Nov 1980||Valleylab, Inc.||Safety monitoring circuit for electrosurgical unit|
|US4232676||16 Nov 1978||11 Nov 1980||Corning Glass Works||Surgical cutting instrument|
|US4237887||23 Jan 1975||9 Dec 1980||Valleylab, Inc.||Electrosurgical device|
|US4281373||16 May 1978||28 Jul 1981||Societe Satelec||High frequency voltage generator|
|US4287557||17 Dec 1979||1 Sep 1981||General Electric Company||Inverter with improved regulation|
|US4296413||28 Sep 1979||20 Oct 1981||General Electric Company||Resistance-bridge to frequency converter with automatic offset correction|
|US4303073||17 Jan 1980||1 Dec 1981||Medical Plastics, Inc.||Electrosurgery safety monitor|
|US4311154||23 Mar 1979||19 Jan 1982||Rca Corporation||Nonsymmetrical bulb applicator for hyperthermic treatment of the body|
|US4314559||12 Dec 1979||9 Feb 1982||Corning Glass Works||Nonstick conductive coating|
|US4321926||16 Apr 1979||30 Mar 1982||Roge Ralph R||Insertion detecting probe and electrolysis system|
|US4334539||28 Apr 1980||15 Jun 1982||Cimarron Instruments, Inc.||Electrosurgical generator control apparatus|
|US4343308||9 Jun 1980||10 Aug 1982||Gross Robert D||Surgical ground detector|
|US6409722 *||6 Jul 1999||25 Jun 2002||Medtronic, Inc.||Apparatus and method for creating, maintaining, and controlling a virtual electrode used for the ablation of tissue|
|US7305311 *||22 Apr 2005||4 Dec 2007||Advanced Energy Industries, Inc.||Arc detection and handling in radio frequency power applications|
|1||Alexander et al., "Magnetic Resonance Image-Directed Stereotactic Neurosurgery: Use of Image Fusion with Computerized Tomography to Enhance Spatial Accuracy" Journal Neurosurgery, 83; (1995) pp. 271-276.|
|2||Anderson et al., "A Numerical Study of Rapid Heating for High Temperature Radio Frequency Hyperthermia" International Journal of Bio-Medical Computing, 35 (1994) pp. 297-307.|
|3||Astrahan, "A Localized Current Field Hyperthermia System for Use with 192-Iridium Interstitial Implants" Medical Physics, 9(3), May/Jun. 1982.|
|4||Bergdahl et al., "Studies on Coagulation and the Development of an Automatic Computerized Bipolar Coagulator" Journal of Neurosurgery 75:1, (Jul. 1991) pp. 148-151.|
|5||Chicharo et al. "A Sliding Goertzel Algorith" Aug. 1996, pp. 283-297 Signal Processing, Elsevier Science Publishers B.V. Amsterdam, NL vol. 52 No. 3.|
|6||Cosman et al., "Methods of Making Nervous System Lesions" In William RH, Rengachary SS (eds): Neurosurgery, New York: McGraw-Hill, vol. 111, (1984), pp. 2490-2499.|
|7||Cosman et al., "Radiofrequency Lesion Generation and Its Effect on Tissue Impedance" Applied Neurophysiology 51 (1988) pp. 230-242.|
|8||Cosman et al., "Theoretical Aspects of Radiofrequency Lesions in the Dorsal Root Entry Zone" Neurosurgery 15:(1984) pp. 945-950.|
|9||Geddes et al., "The Measurement of Physiologic Events by Electrical Impedence" Am. J. MI, Jan. Mar. 1964, pp. 16-27.|
|10||Goldberg et al., "Tissue Ablation with Radiofrequency: Effect of Probe Size, Gauge, Duration, and Temperature on Lesion Volume" Acad Radio (1995) vol. 2, No. 5, pp. 399-404.|
|11||Hadley I C D et al., "Inexpensive Digital Thermometer for Measurements on Semiconductors" International Journal of Electronics; Taylor and Francis. Ltd.; London, GB; Vol. 70, no. 6 Jun. 1, 1991; pp. 1155-1162.|
|12||International Search Report EP 04009964 dated Jul. 13, 2004.|
|13||International Search Report EP 04015981.6 dated Sep. 29, 2004.|
|14||International Search Report EP 05002769.7 dated Jun. 9, 2006.|
|15||International Search Report EP 05014156.3 dated Dec. 28, 2005.|
|16||International Search Report EP 05021944.3 dated Jan. 18, 2006.|
|17||International Search Report EP 05022350.2 dated Jan. 18, 2006.|
|18||International Search Report EP 06000708.5 dated Apr. 21, 2006.|
|19||International Search Report EP 06006717.0 dated Aug. 7, 2006.|
|20||International Search Report EP 06010499.9 dated Jan. 29, 2008.|
|21||International Search Report EP 07001489.9 dated Dec. 20, 2007.|
|22||International Search Report EP 07001491 dated Jun. 6, 2007.|
|23||International Search Report EP 07008207.8; dated Sep. 5, 2007.|
|24||International Search Report EP 07009322.4 dated Jan. 14, 2008.|
|25||International Search Report EP 07010673.7; dated Sep. 24, 2007.|
|26||International Search Report EP 07015601.3 dated Jan. 4, 2008.|
|27||International Search Report EP 07015602.1 dated Dec. 20, 2007.|
|28||International Search Report EP 07019174.7 dated Jan. 29, 2008.|
|29||International Search Report EP 98300964.8 dated Dec. 4, 2000.|
|30||International Search Report EP06022028.2 dated Feb. 5, 2007.|
|31||International Search Report EP06025700.3 dated Apr. 12, 2007.|
|32||International Search Report EP07001481.6 dated Apr. 23, 2007.|
|33||International Search Report EP07001485.7 dated May 15, 2007.|
|34||International Search Report EP07001527.6 dated May 2007.|
|35||International Search Report EP07004355.9 dated May 21, 2007.|
|36||International Search Report EP08004667.5 dated Jun. 3, 2008.|
|37||International Search Report EP08006733.3 dated Jul. 28, 2008.|
|38||International Search Report EP08012503 dated Sep. 19, 2008.|
|39||International Search Report EP08013605 dated Nov. 17, 2008.|
|40||International Search Report EP08015601.1 dated Dec. 5, 2008.|
|41||International Search Report EP08016540.0 dated Feb. 25, 2009.|
|42||International Search Report EP08155780 dated Jan. 19, 2009.|
|43||International Search Report EP08166208.2 dated Dec. 1, 2008.|
|44||International Search Report EPO4707738 dated Jul. 4, 2007.|
|45||International Search Report PCT/US03/33711 dated Jul. 16, 2004.|
|46||International Search Report PCT/US03/33832 dated Jun. 17, 2004.|
|47||International Search Report PCT/US03/37110 dated Jul. 25, 2005.|
|48||International Search Report PCT/US03/37310 dated Aug. 13, 2004.|
|49||International Search Report PCT/US04/02961 dated Aug. 2, 2005.|
|50||International Search Report-Extended EP 06000708.5 dated Aug. 22, 2006.|
|51||Medtrex Brochure "The O.R. Pro 300" 1 p. Sep. 1998.|
|52||Muller et al., "Extended Left Hemicolectomy Using the LigaSure Vessel Sealing System" Innovations That Work; Sep. 1999.|
|53||Ni W et al: "A Signal Processing Method for the Coriolis Mass Flowmeter Based on a Normalized . . . " Journal of Applied Sciences-Yingyong Kexue Xuebao, Shanghai CN, vol. 23 No. 2;(Mar. 2005); 160-164.|
|54||Ogden Goertzel Alternative to the Fourier Transform: Jun. 1993 pp. 485-487 Electronics World; Reed Business Publishing, Sutton, Surrey, BG vol. 99, No. 9. 1687.|
|55||Richard Wolf Medical Instruments Corp. Brochure, "Kleppinger Bipolar Forceps & Bipolar Generator" 3 pp. Jan. 1989.|
|56||Sugita et al., "Bipolar Coagulator with Automatic Thermocontrol" J. Neurosurg., vol. 41, Dec. 1944, pp. 777-779.|
|57||US 6,878,148, 04/2005, Goble et al. (withdrawn)|
|58||Valleylab Brochure "Valleylab Electroshield Monitoring System" 2 pp. Nov. 1995.|
|59||Vallfors et al., "Automatically Controlled Bipolar Electrosoagulation-'COA-COMP'" Neurosurgical Review 7:2-3(1984) pp. 187-190.|
|60||Vallfors et al., "Automatically Controlled Bipolar Electrosoagulation-‘COA-COMP’" Neurosurgical Review 7:2-3(1984) pp. 187-190.|
|61||Wald et al., "Accidental Burns", JAMA, Aug. 16, 1971, vol. 217, No. 7, pp. 916-921.|
|Citing Patent||Filing date||Publication date||Applicant||Title|
|US7761247 *||16 Oct 2007||20 Jul 2010||Advanced Energy Industries, Inc.||Arc detection and handling in radio frequency power applications|
|US8044594||15 Apr 2009||25 Oct 2011||Advanced Energy Industries, Inc.||Power supply ignition system and method|
|US8217299||22 Feb 2007||10 Jul 2012||Advanced Energy Industries, Inc.||Arc recovery without over-voltage for plasma chamber power supplies using a shunt switch|
|US8556890 *||14 Dec 2009||15 Oct 2013||Covidien Ag||Arc based adaptive control system for an electrosurgical unit|
|US8623005||31 Oct 2011||7 Jan 2014||Baylis Medical Company Inc.||Stent graft fenestration|
|US8636730||12 Jul 2010||28 Jan 2014||Covidien Lp||Polarity control of electrosurgical generator|
|US20100094275 *||14 Dec 2009||15 Apr 2010||Covidien Ag||Arc Based Adaptive Control System for an Electrosurgical Unit|
|US20130041355 *||11 Aug 2011||14 Feb 2013||Tammo Heeren||Reducing Damage From A Dielectric Breakdown in Surgical Applications|
|WO2011119933A2 *||25 Mar 2011||29 Sep 2011||Aragon Surgical, Inc.||Impedance mediated control of power delivery for electrosurgery|
|12 Mar 2013||FPAY||Fee payment|
Year of fee payment: 4
|7 Oct 2009||AS||Assignment|
Owner name: COVIDIEN AG, SWITZERLAND
Free format text: CHANGE OF NAME;ASSIGNOR:TYCO HEALTHCARE GROUP AG;REEL/FRAME:023336/0098
Effective date: 20081215
Owner name: COVIDIEN AG, SWITZERLAND
Free format text: CHANGE OF NAME;ASSIGNOR:SHERWOOD SERVICES AG;REEL/FRAME:023336/0074
Effective date: 20070309
Owner name: TYCO HEALTHCARE GROUP AG, SWITZERLAND
Free format text: MERGER;ASSIGNOR:COVIDIEN AG;REEL/FRAME:023336/0087
Effective date: 20081215
|24 Apr 2006||AS||Assignment|
Owner name: SHERWOOD SERVICES AG, SWITZERLAND
Free format text: ASSIGNMENT OF ASSIGNORS INTEREST;ASSIGNOR:WHAM, ROBERT H.;REEL/FRAME:017808/0913
Effective date: 20060424