<< Used Digital Multimeters
<< Hewlett Packard / Agilent Used Test Equipment Price List
/ HP 3478A
HP 3478A provides a low cost, full function, reliable DMM for system measurements. Selectable 3 1/2 to 5 1/2 digit resolution and 5 autoranging functions offer flexibility in automated testing. The 3478A can measure DCV, true RMS ACV, 2- and 4-wire ohms, and dc and ac current. Simple, fast electronic calibration eliminates all adjustments to provide a lower cost of ownership.
Hewlett Packard 3478A DMM Shown
Low Cost of Ownership
The combination of an extremely reliable DMM with features like complete electronic calibration and self-test gives you the lowest cost of ownership. We're so sure of the 3478A's reliability that you can get additional hardware service for less than 2% of the DMMs price per year.
Selectable speed and resolution provide the right capability for your measurement. The 3478A can perform production tests or acquire experimental data at 90 readings/s with 3 1/2 digit resolution, or take 35 readings/s with 130 dB of noise rejection using 4 1/2 digits. The 5 1/2 digit mode offers 100 nVdc and 100 µW resolution for precise measurements. True RMS with 300 kHz bandwidth and 4:1 crest factor provides reliable measurements of ac signals. Fast autoranging makes the first reading useful and accurate.
Designed for Systems
Switchable front/rear inputs permit flexible system connections. The Voltmeter Complete output and External Trigger input allow synchronization of the 3478A with a scanner for fast multiplexed measurements without the delay of software commands. The test program can write prompt messages or results on the alphanumeric display. The operator can respond by pressing the 3478A's SRQ key to interrupt the controller and start the next test. Built-in self-test capability assures proper operation.
Complete calibration of the 3478A is accomplished without any internal adjustment or removing the instrument's covers. The simple and fast electronic calibration procedure lowers the cost of ownership. You only need to connect standards to the 3478A and store calibration constants in the 3478A's non-volatile memory. Calibration can be done manually from the front panel or automatically using HP-IB.
|3478A||Agilent / HP 3478A Digital Multimeter, Refurbished||$325.00|
|315||100V, 50Hz||Please Call|
|901||Front Handle Kit|
|908||Rack Mount Kit|